Reliability and criterion validity of two applications of the iPhone™ to measure cervical range of motion in healthy participants
© Tousignant-Laflamme et al.; licensee BioMed Central Ltd. 2013
Received: 4 July 2012
Accepted: 14 June 2013
Published: 5 July 2013
Summary of background data
Recent smartphones, such as the iPhone, are often equipped with an accelerometer and magnetometer, which, through software applications, can perform various inclinometric functions. Although these applications are intended for recreational use, they have the potential to measure and quantify range of motion. The purpose of this study was to estimate the intra and inter-rater reliability as well as the criterion validity of the clinometer and compass applications of the iPhone in the assessment cervical range of motion in healthy participants.
The sample consisted of 28 healthy participants. Two examiners measured cervical range of motion of each participant twice using the iPhone (for the estimation of intra and inter-reliability) and once with the CROM (for the estimation of criterion validity). Estimates of reliability and validity were then established using the intraclass correlation coefficient (ICC).
We observed a moderate intra-rater reliability for each movement (ICC = 0.65-0.85) but a poor inter-rater reliability (ICC < 0.60). For the criterion validity, the ICCs are moderate (>0.50) to good (>0.65) for movements of flexion, extension, lateral flexions and right rotation, but poor (<0.50) for the movement left rotation.
We found good intra-rater reliability and lower inter-rater reliability. When compared to the gold standard, these applications showed moderate to good validity. However, before using the iPhone as an outcome measure in clinical settings, studies should be done on patients presenting with cervical problems.
KeywordsRange of motion Outcome measures Reliability Criterion validity iPhone Cervical spine Digital inclinometer
Cervical disorders are major health problems in our society and an important source of disability . The mean prevalence of neck pain in the general population is 23.1% with a higher incidence noted in office and computer workers . It is also one of the most common reasons to visit a health care professional . Consequences of cervical disorders are multiple and include deficits such as pain and decreased range of motion (ROM) , which may reduce social participation and even lead to a sick leave .
Assessment of ROM is a significant part of the physical therapist’s role when evaluating a patient presenting with cervical disorders. Indeed, it helps to establish the clinical diagnosis and the prognosis, and also helps to elaborate an individualized treatment plan . ROM is also an objective measure, which is essential to monitor the patient’s evolution throughout therapy. For these reasons, valid and reliable assessment tools are necessary.
Numerous valid tools are currently available to measure cervical ROM: they include inclinometers [6–8], digital inclinometers [9–13], measuring tape and goniometer , and the Cervical Range of Motion Device (CROM) [4, 5, 8, 10, 14–20]. The CROM is one of the most used tools among clinicians . The systematic review of Williams et al. showed that the CROM has a good reliability for all cervical spine movements (ICCs = 0,58-0,99) and validity when compared to a gold standard (X-ray) (ICCs = 0,82-0,98). However, it is a relatively expensive instrument and only useful for the assessment of the cervical spine.
Regarding the digital inclinometers, only a few validity and reliability studies have been realized for cervical spine ROM [6–8]. Among digital inclinometers, the EDI-320 is an instrument that demonstrated good psychometric qualities (reliability ICCs = 0,69-0,96) according to two studies [4, 18]. Unfortunately, this instrument is no longer available on the market. More recently, Prushansky et al.  showed that the conventional digital inclinometer has a good to excellent reliability (ICCs = 0,82-0,94). It also has good validity (ICCs = 0,62-0,83) when compared to the ultrasonography-based Zebris CMS 70P (Zebris Medizintechnik Gmbh, Isny, Germany). These authors measured cervical spine movements in three planes of movement and showed good results (ICCs = 0,82-0,94) with rotations in supine.
Recent smartphones are often equipped with an accelerometer (gravity sensor) and magnetometer (digital compass), which, through software applications, can perform various inclinometric functions. These applications are intended for recreational use, but have the potential to measure and quantify range of motion in many articulations, such as the cervical spine. For instance, previous studies have demonstrated the potential use of some applications in rehabilitation [23, 24] and in ROM measurement . The iPhone is easy to use and requires minimum training. Moreover, this instrument could allow the examiner (therapist) to obtain valid cervical ROM measurements, which can detect deficits in cervical ROM. Considering potential use of smartphones in rehabilitation and the favourable results obtained with digital inclinometers , the current study proposes to examine the psychometric properties of two applications (clinometer and compass) of the iPhone. The specific objectives are to determine the intra and inter-rater reliability of these two applications in the assessment of cervical ROM, as well as the criterion validity using CROM as the gold standard.
Design of the study
In this study, we used a descriptive correlational design to determine the reliability of the iPhone using intra and inter-rater reliability. For exploring the validity of these applications, we used criterion validity using the CROM as the gold standard. Because of the absence of any study on the reliability or validity of the iPhone for the measurement of cervical ROM, the population used in this study is composed of healthy participants (without neck pain and/or ROM deficits).
Our sample consisted of 28 healthy volunteers (9 men and 19 women) aged from 19 to 43 years old (mean ± SD: 23 ± 6). Participants were included if they were 18 years of age or older and had neither cervical spine problem or neck pain. We excluded persons with cervical pathology (ex. painful diagnosis of arthritis or whiplash during the past year), psychiatric condition (ex. dementia, amnesia, delirium) or neurological disease (ex. Multiple sclerosis, Lou Gehrig’s Disease). The population included in this protocol was a convenient sample, recruited by purposive and snowball sampling. All volunteers consented for their participation in the study and did not receive monetary rewards or compensation for their time and participation to this study. The study was conducted in accordance with the Helsinki Declaration after approval from the ethics review board of the Centre hospitalier universitaire de Sherbrooke (project #10-199). All participants read the protocol, and a written consent was obtained in agreement with local ethics guidelines’. The study took place at the School of Rehabilitation of the Université de Sherbrooke. Considering the novel aspect related to the use of smartphones to measure ROM and the fact that we wanted to explore the psychometric properties of these applications, we opted for the recruitment of healthy subjects. All were assessed by the same instruments and the same observers.
The iPhone is a smartphone with many possible applications. The application used to measure the cervical ROM in frontal and sagittal planes is Clinometer (Peter Breitling, Version 3.3, http://www.plaincode.com/products), an application designed using the three inbuilt accelerometers (LIS302DL accelerometer). This application uses the internal three axes linear accelerometer to measure the direction of gravity’s pull. For this, the gyroscope stays in one position, no matter the orientation. When placed against a solid surface, the inclinometer compares the angle of the object to the gyroscope, and displays the results using the software interface.
Cervical Range of Motion Device (CROM)
The CROM was used for the measurement of cervical flexion, extension, lateral flexions and rotations. This eyeglasses-like instrument has three inclinometers placed at three different positions: one near the left ear for flexion/extension (sagittal plan) and another for the lateral flexions on forehead (frontal plane) and both are gravity dependent. Finally, the one on the top of the head (horizontal plane) is used for the measurement of rotations; it is magnetic dependant, therefore, a magnetic brace must be placed around the neck. This instrument was used as our gold standard considering that its reliability and validity have been studied extensively [10, 18, 26, 27].
For the purpose of this study, participants were simply asked to perform maximal (end-range) neck flexion, extension, left side flexion, right side flexion, left head rotation and right head rotation. Each participant was asked to perform neck movement at his/her own pace without going to fast.
Selection of examiners for the reliability study
Four students in physical therapy received three hours of training to adequately manipulate the CROM device. In their training, they also taught other classmates how to use the device during a two-hour session to enhance their competence in using the CROM. Following their training session, they determined which anatomical point of reference should be used with the iPhone and they trained for an hour to make sure their method was standardized. They then measured their own cervical ROM with the CROM and the iPhone (each student was measured twice). The intra-rater reliability was calculated for each student with the intraclass correlation coefficient (ICC). The two students with higher ICC’s results (ICC = 0,79 and 0,81) were assigned as examiner for the reliability part of the study. These two practiced their techniques in another two-hour session with four volunteers to standardize the procedure. Overall, the examiners had eight hours of training with the two instruments. This was done in order to minimize the error originating from examiners.
Selection of the examiner for the validity study
Between the two examiners, the one with the highest intra-rater reliability (highest ICCs) was chosen to undertake the validity study. This was done in order to minimize the error originating from examiners.
During all data collection sessions, the participants were instructed on the procedures. They were then asked to warm-up with five repetitions of all cervical movements. Afterwards, stabilizing straps were installed to prevent any trunk and shoulder movements during the movement’s execution (the same procedure were used during the selection of the examiners).
All measures were taken in the same order: flexion, extension, right and left lateral flexions and right then left rotations. This was done in order to minimize the possible bias induced by thixotropy .
For the sagittal and frontal plans, measures always corresponded to the total range (in degrees): the difference between final and initial measure. For example, a starting position of 5° slightly in extension and an end-range of 65° in flexion give us a total flexion of 70° (65° - (−5°) = 70°). Although the procedures for the CROM indicate to only take the final measure (angle at end-range), we could not use this method for the iPhone since our landmark was not necessary at 0° (i.e.: iPhone aligned with the ear), whereas it is always at 0° for the CROM. The total range (in degrees) was also used for the rotations movements.
Procedures for the reliability study
Two examiners entered two different rooms with a paired observer. They took all cervical ROM measures (flexion, extension, right then left lateral flexions and right then left rotations) with the iPhone while their paired observer wrote down the measures.
Procedures for the validity study
Examiner 1 (which had the highest intra-rater ICCs during the selection of examiners process) entered room 1 with his paired observer. He took cervical ROM measures (flexion, extension, right then left lateral flexions and right then left rotations) with the CROM. The examiner then changed room in clockwise motion until he had taken a measure of each movement for each participant with the CROM; this allowed us to establish the criterion validity of the iPhone in comparison to this gold standard.
Sample size calculation
In order to have minimal significant ICC value of 0.60 (1-β = 0.80; α = 0.05), a minimum of 20 subjects was required.
For the reliability part of the study, intra and inter-rater reliability were estimated with the intraclass correlation coefficient (ICC). ICC is a statistic designed to measure the size and direction of the association between two variables . The values vary between −1 (perfect negative association) and +1 (perfect positive association). Different guidelines exist for the interpretation of ICC, but one reasonable scale is that an ICC value of less than 0,4 indicates poor reproducibility; ICC values in the range of 0,4 to 0,75 indicate fair to good reproducibility, and an ICC value of greater than 0,75 shows excellent reproducibility .
Reference values of the interpretation of ICCs
Intra-rater reliability and descriptive statistics for measurements of cervical spine ROM using the iPhone
Examiner 1 / iPhone 4
Examiner 2 / iPhone 3 GS
Trial 1 (Mean ± SD)
Trial 2 (Mean ± SD)
95% CI for ICC
Trial 1 (Mean ± SD)
Trial 2 (Mean ± SD)
95% CI for ICC
55,3° ± 9,9°
56,6° ± 8,2°
58,9° ± 8,1°
61,4° ± 8,0°
83,3° ± 15,2°
81,1° ± 14,6°
90,7° ± 15,9°
84,3° ± 13,2°
Right lateral flexion
44,6° ± 8,3°
46,5° ± 8,7°
52,0° ± 8,1°
41,5° ± 7,2°
Left lateral flexion
46,6° ± 7,1°
48,3° ± 7,6°
50,0° ± 6,2°
49,9° ± 6,7°
71,1° ± 15,8°
73,6° ± 13,4°
87,8° ± 31,0°
91,7° ± 27,6°
74,5° ± 16,2°
76,2° ± 13,1°
87,6° ± 29,0°
76,3° ± 24,5°
Descriptive and inter-rater reliability for measurements of cervical spine ROM using the iPhone
Examiner 1 (mean ± SD)
Examiner 2 (mean ± SD)
56,0° ± 8,3°
60,1° ± 7,0°
82,2° ± 14,0°
87,5° ± 12,0°
Right lateral flexion
45,6° ± 7,8°
51,8° ± 7,0°
Left lateral flexion
47,5° ± 6,8°
50,0° ± 5,8°
72,4° ± 14,5°
89,7° ± 21,3°
75,4° ± 14,7°
82,0° ± 22,3°
When compared to our chosen gold standard (CROM), we observed good validity for the movements of flexion (ICC = 0,76; r = 0.69, p = 0.001), right lateral flexion (ICC = 0,85; r = 0.80, p < 0.001) and left lateral flexion (ICC = 0,70; r = 0.63, p < 0.001).
Criterion validity of the iPhone compared to the CROM
iPhone (Mean ± SD)
CROM (Mean ± SD)
ICC ± [95% CI]
Pearson’s r (p-value)
56.0° ± 8.3°
57.1° ± 8.4°
0.69 p < 0.001
82.2° ± 14.0°
85.5° ± 12.3°
0.56 p = 0.002
Right lateral flexion
45.6° ± 7.8°
44.7° ± 8.2°
0.80 p < 0.001
Left lateral flexion
47.5° ± 6.8°
47.2° ± 7.0°
0.63 p < 0.001
72.4° ± 14.5°
73.8° ± 8.6°
0.58 P < 0.01
75.4° ± 14.7°
74.9° ± 8.6°
0.38 p = 0.04
This study is the first to examine the predictive value of two applications of the iPhone, which have the capability to measure cervical ROM using the CROM as the accepted gold standard. Although a few studies were already done on the validity of a digital device for the measurement of cervical ROM, no previous study was done on the digital inclinometer and/or the compass of the iPhone for the measurement of cervical ROM.
Reliability estimates are very important psychometric properties since before an instrument can be considered valid, it needs to be reliable. The findings of this study showed that when the cervical ROM is measured with the iPhone by the same examiner (intra-rater reliability), similar results can be expected from one session to the next. It is possible to compare the current results of intra-rater reliability with digitals inclinometers such as Electronic Digital goniometer (EDI-320) for the measurement of active neck movement. For instance, in the sagittal plane, we found good reliability for flexion (ICC = 0,78; 95% CI: 0,58-0,89) and extension (ICC = 0,84; 95% CI: 0,69-0,92). Tousignant et al.  also reported good reliability for flexion (ICC = 0.77; 95% CI: 0,62-0,87) and extension (ICC = 0,83; 95% CI: 0,63-0,92) using the EDI-320. However, this device is not on the market anymore. In comparison, the reliability coefficients found in our study were slightly higher.
For the frontal plane, we found good intra-rater reliability (ICCs = 0.77-0.78; 95% CI: 0,56-0,89), while Prushansky et al.  found similar results with a digital inclinometer (ICCs = 0.82-0.90; 95% CI: 0,61-0,95).
Finally, in the transverse axis, we found moderate to good intra-rater reliability (ICCs = 0.66-0.74; 95% CI: 0,39-0,87), while Prushansky et al.  observed higher ICCs (ICCs = 0.84-0.92; 95% CI: 0,68-0,96). This might be explained by the fact that they took their measurements with the inclinometer while the subjects were in supine position, whereas we used the compass rather than the inclinometer of the iPhone. Since the compass is not influenced by gravity, but rather by orientation of the iPhones, it has more potential source of error than the inclinometer, which could have easily influenced the intra and inter-rater ICCs. Furthermore, the magnetometer which serves as the hardware for the compass application is more sensible of the presence of electro-magnetic fields which is another factor that could have contribute to the lower ICCs for the measurements of neck rotation.
When the ROM measured by two independent examiners were compared, our ICCs were moderate for movements in the sagittal plane (ICCs = 0.48-0,49; 95% CI: 0,14-0,72) and in the frontal plan (ICCs = 0.40-0,54; 95% CI: 0,04-0,75). When we look closely at our results, we found that examiner 2, who used an iPhone generation 3GS, always had higher ROM measures than examiner 1, who used an iPhone generation 4. Considering that Apple uses an LIS302DL accelerometer for both iPhones 4 and 3GS and the two different generations of iPhone had the same operating system (iOS 4), factors related to the positioning of the iPhone might explain this observation. We also found poor correlation in transverse plan (ICC = 0.07-0,09; 95% CI: -0,30-0,44), which again might be explained by the presence of electro-magnetic fields that could influence the measure. On the other hand, it could also be attributed to the examiner since examiner 2 showed lower intra-rater reliability.
Cervical ROM measured with the iPhone presented comparable results (moderate to good validity) when compared to the ROM measured with the CROM for all cervical movements, except for the movement of left rotation (ICC = 0.43). On the basis of this relation, the validity of the iPhone can be considered good for these movements for a same examiner, except for rotation. The poor results observed for the movements of rotation (ICC < 0.60) may partly be explained by the fact that it was measured by an application very sensible to electro-magnetic fields. This can lessen the accuracy of the, measurement. It could also be explained by the movement and/or positioning of the iPhone during the measurement of cervical rotation.
To our knowledge, no study examining the validity of the iPhone for assessing cervical ROM has been published. However, a recent article on the reliability and validity of a relatively inexpensive digital inclinometer reported results that were similar to our findings in sagittal and frontal planes: a good reliability (ICCs = 0,82-0,94) but lower validity (r = 0,62-0,83). Results were different for the rotation movements: a good reliability (ICCs = 0,84-0,92) and poor validity (results not reported). Their better results obtained for the reliability of rotations might be explain by the fact that rotations measured in supine position .
Our results show that measures of extension and right rotation had poor inter-rater reliability and thus mined the validity of this measure. This discrepancy may be attributed to the data collection procedures or the placement of the iPhone on the top of participant’s head. Special efforts were made in this study to minimize this type of error, but we suggest that future measurements of rotation movements might be done with the iPhone on the top of the forehead while the person is lying supine as done by Prushansky .
Strengths and limitations
First, the two examiner’s initial preparation (training) with the CROM represents strength. The assessment of the examiner’s skills showed that they were competent (ICC > 0,65) in the use of the method and the device (E1: ICC = 0,81; 95% 0,56-0,92. E2: ICC = 0,79; 95% 0,52-91) (see Table 1 for ICC reference values). For the validity study, we purposely chose examiner 1 in order to minimize the source of error coming from the examiner.
Second, standardization of the procedures also helped minimize random errors. To achieve this, all participants were stabilized in order to avoid compensation. Also, the research assistant always gave the same instructions before each measurement for all participants and the environment was identical during all the data collection process: same rooms, same orientation or the participants (facing east), same chairs, etc.
Thirdly, measures were taken with the iPhone and the CROM were always taken in the same order. Thus, if the cervical ROM increased with repetitions, the pattern would be the same for all participants and would not influence our results.
Finally, the iPhone measures were always taken before the CROM measure to prevent an information bias. Due to the numerous measurements took with the iPhone, we considered that it would have been impossible for the examiner to remember all the results and influence its readings using the CROM. Therefore, we think that this help minimized an information bias.
This study also had limitations. First, data was collected on a sample of healthy participants, which limits the external validity. Although we tried to minimize bias affecting the internal validity, but the fact that examiner 1 had higher intra-rater ICCs than examiner 2 might partly explain the modest results for the inter-rater reliability.
Implications of this study relate to the use of the iPhone to measure the cervical ROM in patients without neck dysfunction. The iPhone is a popular device and has good potential for clinical use. This instrument is easy to use and requires minimum training. Moreover, this instrument could allow the examiner (therapist) to obtain valid cervical ROM measurements, which can detect deficits in cervical ROM.
In the current non-probabilistic sample of healthy participants, we found that the iPhone had good intra-rater reliability but lower inter-rater reliability. When compared to a gold standard (CROM), the iPhone showed moderate to good validity for movements in the sagittal and frontal plans, but poor validity for rotation movements. At this stage, we cannot recommend the use of the iPhone to measure cervical range of motion in all directions. Moreover, before using the iPhone as an outcome measure in clinical settings, we should focus on finding better positioning method for the measurement of cervical rotation and more importantly, studies should be done on patients presenting with cervical problems.
At the time of data collection, NB, AD and CAV are students were final year physical therapy students at the Université de Sherbrooke. YTL is an associate professor at the School of rehabilitation of the Université de Sherbrooke and is a supported member of the Centre de recherche Clinique Etienne-LeBel du CHUS.
The authors gratefully acknowledge the invaluable assistance of Nathalie Carrier, the participating statistician of the Sherbrooke University Hospital Center, Ruknodine Qurbani and Noémie Royer who helped us in the beginning, and all subjects who volunteered to participate in this study. We would also like to thanks Professor Philippe Archambault (OT, PhD) for his helpful comments and revision of the manuscript.
- Dvir Z, Prushansky T: Reproducibility and instrument validity of a new ultrasonography-based system for measuring cervical spine kinematics. Clin Biomech. 2000, 15 (9): 658-664. 10.1016/S0268-0033(00)00033-4.View Article
- Hoy DG, Protani M, De R, Buchbinder R: The epidemiology of neck pain. Best Pract Res Clin Rheumatol. 2010, 24 (6): 783-792. 10.1016/j.berh.2011.01.019.View ArticlePubMed
- Buskila D, Sarzi-Puttini P, Ablin JN: The genetics of fibromyalgia syndrome. Pharmacogenomics. 2007, 8 (1): 67-74. 10.2217/146224220.127.116.11.View ArticlePubMed
- Lee H, Nicholson LL, Adams RD: Cervical range of motion associations with subclinical neck pain. Spine. 2004, 29 (1): 33-40. 10.1097/01.BRS.0000103944.10408.BA.View ArticlePubMed
- Reese NB, Bandy WD: Joint Range of motion and muscle lenght testing. 2002, St-Louis, Missouri: Saunders Elservier
- Bush KW, Collins N, Portman L, Tillett N: Validity and intertester reliability of cervical range of motion using inclinometer measurements. The Journal of Manual & Manipulative Therapy. 2000, 8 (2): 52-61. 10.1179/106698100790819546.View Article
- Tucci SM, Hicks JE, Gross EG, Campbell W, Danoff J: Cervical motion assessment: a new, simple and accurate method. Arch Phys Med Rehabil. 1986, 67 (4): 225-230.PubMed
- Hole DE, Cook JM, Bolton JE: Reliability and concurrent validity of two instruments for measuring cervical range of motion: effects of age and gender. Man Ther. 1995, 1 (1): 36-42. 10.1054/math.1995.0248.View ArticlePubMed
- Mayer T, Brady S, Bovasso E, Pope P, Gatchel RJ: Noninvasive measurement of cervical tri-planar motion in normal subjects. Spine. 1993, 18 (15): 2191-2195. 10.1097/00007632-199311000-00007.View ArticlePubMed
- Love S, Gringmuth RH, Kazemi M, Cornacchia P, Schmolke M: Interexaminer and intraexaminer reliability of cervical passive range of motion using the CROM and Cybex 320 EDI. J Can Chiropr Assoc. 1998, 42 (4): 222-228.PubMed Central
- Hoving JL, Pool JJ, Mameren H, Deville WJ, Assendelft WJ, Vet HC, Winter AF, Koes BW, Bouter LM: Reproducibility of cervical range of motion in patients with neck pain. BMC Musculoskelet Disord. 2005, 6: 59-10.1186/1471-2474-6-59.PubMed CentralView ArticlePubMed
- Zwart JA: Neck mobility in different headache disorders. Headache. 1997, 37 (1): 6-11. 10.1046/j.1526-4610.1997.3701006.x.View ArticlePubMed
- Tousignant M, Boucher N, Bourbonnais J, Gravelle T, Quesnel M, Brosseau L: Intratester and intertester reliability of the cybex electronic digital inclinometer (EDI-320) for measurement of active neck flexion and extension in healthy subjects. Man Ther. 2001, 6 (4): 235-241. 10.1054/math.2001.0419.View ArticlePubMed
- Nilsson N, Christensen HW, Hartvigsen J: The interexaminer reliability of measuring passive cervical range of motion, revisited. J Manip Physiol Ther. 1996, 19 (5): 302-305.
- Olson SL, O’Connor DP, Birmingham G, Broman P, Herrera L: Tender point sensitivity, range of motion, and perceived disability in subjects with neck pain. J Orthop Sports Phys Ther. 2000, 30 (1): 13-20.View ArticlePubMed
- Peolsson A, Hedlund R, Ertzgaard S, Oberg B: Intra- and inter-tester reliability and range of motion of the neck. Physiother Can. 2000, 52 (3): 233-242.
- Rheault W, Albright B, Beyers C, Franta M, Johnson A, Skowronek M, Dougherty J: Intertester reliability of the cervical range of motion device. J Orthop Sports Phys Ther. 1992, 15 (3): 147-150.View ArticlePubMed
- Tousignant M, Ld B, O’Donoughue S, Grahovac S: Criterion validity of the cervical range of motion (CROM) goniometer for cervical flexion and extension. Spine. 2000, 25 (3): 324-330. 10.1097/00007632-200002010-00011.View ArticlePubMed
- Youdas JW, Carey JR, Garrett TR: Reliability of measurements of cervical spine range of motion--comparison of three methods. Phys Ther. 1991, 71 (2): 98-104.PubMed
- Youdas JW, Garrett TR, Suman VJ, Bogard CL, Hallman HO, Carey JR: Normal range of motion of the cervical spine: an initial goniometric study. Phys Ther. 1992, 72 (11): 770-780.PubMed
- Williams MA, McCarthy CJ, Chorti A, Cooke MW, Gates S: A systematic review of reliability and validity studies of methods for measuring active and passive cervical range of motion. J Manip Physiol Ther. 2010, 33 (2): 138-155. 10.1016/j.jmpt.2009.12.009.View Article
- Prushansky T, Deryi O, Jabarreen B: Reproducibility and validity of digital inclinometry for measuring cervical range of motion in normal subjects. Physiotherapy Research International. The Journal For Researchers And Clinicians In Physical Therapy. 2010, 15 (1): 42-48.
- Lee BC, Kim J, Chen S, Sienko KH: Cell phone based balance trainer. J Neuroeng Rehabil. 2012, 9: 10-10.1186/1743-0003-9-10. -0003-9-10PubMed CentralView ArticlePubMed
- Wu HH, Lemaire ED, Baddour N: Conference proceedings: Annual International Conference of the IEEE Engineering in Medicine and Biology Society. IEEE Engineering in Medicine and Biology Society.Conference. Change-of-state determination to recognize mobility activities using a BlackBerry smartphone. 2011, 5252-5255.
- Shin SH, du Ro H, Lee OS, Oh JH, Kim SH: Within-day reliability of shoulder range of motion measurement with a smartphone. Man Ther. 2012, 17 (4): 298-304. 10.1016/j.math.2012.02.010.View ArticlePubMed
- Audette I, Dumas J, JC N, SDS J: Validity and between-day reliability of the cervical range of motion (CROM) device. J Orthop Sports Phys Ther. 2010, 40 (5): 318-323.View ArticlePubMed
- Tousignant M, Smeesters C, Breton A, Breton Ã, Corriveau H: Criterion validity study of the cervical range of motion (CROM) device for rotational range of motion on healthy adults. J Orthop Sports Phys Ther. 2006, 36 (4): 242-248.View ArticlePubMed
- Lieber RL, Bodine-Fowler SC: Skeletal muscle mechanics: implications for rehabilitation. Phys Ther. 1993, 73 (12): 844-856.PubMed
- Polgar S, Thomas SA: Introduction to research in the health science. 2008, Philadelphia: Churchill-Livingstone Elsevier, fifth edition
- Fleiss JL: Design and analysis of clinical experiments. 1999, New York: Willey Classical LibraryView Article
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.