Variability in bimanual wheelchair propulsion: consistency of two instrumented wheels during handrim wheelchair propulsion on a motor driven treadmill
© Vegter et al.; licensee BioMed Central Ltd. 2013
Received: 26 January 2012
Accepted: 24 January 2013
Published: 29 January 2013
Handrim wheelchair propulsion is a complex bimanual motor task. The bimanually applied forces on the rims determine the speed and direction of locomotion. Measurements of forces and torques on the handrim are important to study status and change of propulsion technique (and consequently mechanical strain) due to processes of learning, training or the wheelchair configuration. The purpose of this study was to compare the simultaneous outcomes of two different measurement-wheels attached to the different sides of the wheelchair, to determine measurement consistency within and between these wheels given the expected inter- and intra-limb variability as a consequence of motor control.
Nine able-bodied subjects received a three-week low-intensity handrim wheelchair practice intervention. They then performed three four-minute trials of wheelchair propulsion in an instrumented hand rim wheelchair on a motor-driven treadmill at a fixed belt speed. The two measurement-wheels on each side of the wheelchair measured forces and torques of one of the two upper limbs, which simultaneously perform the push action over time. The resulting data were compared as direct output using cross-correlation on the torque around the wheel-axle. Calculated push characteristics such as power production and speed were compared using an intra-class correlation.
Measured torque around the wheel axle of the two measurement-wheels had a high average cross-correlation of 0.98 (std=0.01). Unilateral mean power output over a minute was found to have an intra-class correlation of 0.89 between the wheels. Although the difference over the pushes between left and right power output had a high variability, the mean difference between the measurement-wheels was low at 0.03 W (std=1.60). Other push characteristics showed even higher ICC’s (>0.9).
A good agreement between both measurement-wheels was found at the level of the power output. This indicates a high comparability of the measurement-wheels for the different propulsion parameters. Data from both wheels seem suitable to be used together or interchangeably in experiments on motor control and wheelchair propulsion performance. A high variability in forces and timing between the left and right side were found during the execution of this bimanual task, reflecting the human motor control process.
Keywords(MeSH) Wheelchairs Rehabilitation Biomechanics Motor skills
Handrim wheelchair propulsion is the means of ambulation for a large group of people with a disability. However, handrim wheelchair propulsion is straining and (overuse) injuries to the upper extremities, e.g. shoulder pain or carpal tunnel syndrome, among wheelchair-dependent persons are common [1–3]. Therefore, a better understanding of wheelchair skill, physical capacity and the impact of wheelchair mechanics and fitting on performance are important [4–6]. Research over the past 30 years has led to a number of studies on the physiology and biomechanics of wheeled mobility [7, 8]. Due to the complexity of instrumentation this research was primarily lab-based. Only more recently ambulant instrumentation for both physiological and biomechanical outcomes became available, which today even evolved into commercially available clinical tools [9, 10].
Measurements of forces and torques on the handrim of a wheelchair are important to study change of propulsion technique due to learning, training or the effect of changes to the wheelchair. From a scientific point of view this provides a deeper understanding of the universal principles regarding the motor control of wheelchair propulsion, while from a clinical perspective it can help to better tailor the properties of a wheelchair to a patients needs’ and develop intervention protocols with respect to propulsion technique and strategy . Over time, different studies have used different ways to instrument the wheels to gain insight in the forces and timing involved in wheelchair propulsion, varying from instrumented ergometers to specialized wheels [9, 10, 12–16]. These measurement systems have been used to describe unilaterally the cyclic nature of handrim propulsion analogous to gait analysis. For example, frequency of pushes, peak forces and torques and the wheel angle covered within a push have been used to describe the motor learning process of novel wheelchair users . Besides propulsion technique these wheels are able to measure the power output of the wheelchair-user combination, making it possible to calculate mechanical efficiency when combined with cardio-respiratory measurements and energy calculations .
Most studies on propulsion technique measured this essentially bimanual propulsion task unilaterally and focused on the description of propulsion characteristics in dependence of a variety of different interventions. Yet, due to both internal control processes and external perturbations interlimb variation is expected . Studying unilateral wheelchair propulsive mechanics provide biomechanical information about propulsion technique measures like peak forces and push time. However, wheelchair propulsion is a bimanual task, and studies in the area of bimanual motor control have shown that the limbs are not controlled independently, but are coupled to each other. This implies that principles of interlimb coordination cannot be derived from the study of single-limb movements .
Only few studies addressed bimanual upper limb consistency or the variability for that matter of motor performance in this task by using two instrumented wheels simultaneously [16, 21, 22]. Moreover, the provided data on reliability or validity of these measurement-wheels or related ergometer technology in literature are very scarce [10, 23–25], let alone the comparability of different measurement-wheels.
One way to compare the measurement-wheels is to fit them to both sides of the same wheelchair, during a steady-state propulsion task on a treadmill. Although both wheels examine the same performance of wheelchair propulsion, the motor control process involves a combined movement of the two upper limbs, which determines speed and direction. When comparing the pushes on both wheels, one faces the problem that human movement is intrinsically variable, both within and between individuals . At present, there is a growing recognition that this variability (e.g. intra- and interlimb variability) is not simply the reflection of noise, but contains features that provide insight about normal learning and pathological processes [27–29]. Consequently for comparison of the wheels it is necessary to isolate the consistency of measurement from the variability inherent to the motor control process. Therefore a time dependent one-on-one comparison between pushes of the left and right arm are not expected to provide a suitable outcome measure for comparing the measurement-wheels against each other. Namely at this relatively small time-scale variability due to motor control and task variability, despite propelling at a constant speed on a motor driven treadmill, is to be expected. Indeed, one study using two Smartwheels specifically looked at the asymmetry of wheelchair propulsion and showed side-to-side differences when matching three pushes left and right .
Yet, the set task of straightforward steady-state propulsion on a level treadmill should be intrinsically stable over a larger time-scale and should lead to comparable mean outcome values for the left and right side, resulting in a constant mean power output (product of torque and angular velocity) over time. In the current study it is therefore assumed that systematic differences in unilateral mean power output between both wheels, when propelling at constant speed on a motor driven treadmill, should indicate differences in measurement systems rather then motor variability. For instance van der Woude et al. showed a high correlation (r=0.97) of left- and right-hand sprint power averaged over 30 seconds for 67 wheelchair athletes on a computer controlled wheelchair simulator . Although steering on this ergometer is not critical like on a motor driven treadmill, this high interlimb consistency in power production still exists.
Traditionally power output on a motor-driven treadmill is determined through a separate drag test . In the current study, the drag test, combined with the use of a pulley system is used to impose an additional drag force of known magnitude to the wheelchair-user combination on the treadmill . The outcomes of the measurement wheels are compared with this other form of measuring power output.
Specifically we studied 1) if the two measurement-wheels (Optipushb, Smartwheela) provided comparable time-averaged data for the left and right hand side during steady-state wheelchair propulsion on a motor-driven treadmill in a group of trained able-bodied subjects and 2) if the power output values for the measurement-wheels were comparable to the power output based on a separate drag test.
Answering these questions will enable researchers and practitioners to better interpret results from both measurement-wheels published in previous studies, and use both wheels in the same evaluation setup in the future. Furthermore it gives information on how earlier estimations of power output, using a drag test and a pulley system, compare to the outcomes of the measurement-wheels. Finally this study will help to further our understanding of details of bimanual variability in propulsion technique during steady-state handrim wheelchair propulsion.
After having given written informed consent, 9 able-bodied subjects participated in the study. Criteria for inclusion were male, between 18–65 years, no prior experience in wheelchair propulsion, and absence of any medical contra-indications. To compare with earlier research in our laboratory only male subjects were selected. The study was performed according to the guidelines of the Ethics Committee of the Faculty of Human Movement Sciences, VU University Amsterdam (ECB 2011–46).
The regular rear 24 inch wheels of the standardized wheelchair were replaced with two instrumented wheels; on the left the OptiPushb (6.0 kg, Max Mobility) and on the right the Smartwheela (4.9 kg, 3-Rivers). Both wheels measure 3-dimensional forces and torques applied to the handrim, combined with the angle under which the wheel is rotated. These variables were the only ones used in this experiment for data processing; further data processing and interpretation as done by the respective software packages was not included in the current study. Data were wirelessly transferred to a laptop at 200 Hz (Optipush) and 240 Hz (Smartwheel). Both wheels were synchronised by an electronic pulse at the start of each measurement.
Push time (s)
Time from the start of positive torque to the stop of positive torque for an individual push.
tend(i) - tstart(i)
Cycle time (s)
Time from the start of positive torque to the next start of positive torque.
tend(i) − tend(i − 1)
Contact angle (rad)
Angle at the end of a push minus the angle at the start.
Øend(i) − Østart(i)
3d peak force during the push phase
Maxstart : end(Fx2 + Fy2 + Fz2)0.5
Mean Power/push (W)
The mean power during the push phase.
Meanstart : end(Tz • Δ Ø)
The power integrated over the duration of the push.
∑start : end(Tz • Δ Ø)
Pushes per minute.
Mean power/minute (W)
The mean total (unilateral) power (Tz*Angular velocity) during a complete series of cycles in a minute multiplied by 2.
2 • (∑start(i) : start(n)(Tz • Δ Ø))
First, a cross-correlation was performed between the torque signals around the axle of each of the wheels for each subject, over the whole last minute of each practice block. We were specifically interested in correlation and the time lag between the two measurement-wheels. Possible differences in correlation and time lag were evaluated with repeated-measures Anova.
Secondly, the different biomechanical variables averaged over a minute were compared between the measurement-wheels with an intra-class correlation (ICC) over the different trials. A case 3 ICC was used to compare the degree of absolute agreement of the measurements that are the averages of the three independent 4-minute blocks with the two measurement-wheels as fixed judges . A case 1 ICC was also performed over the three 4-minute blocks within each wheel to relate these within wheel outcomes to the between wheel outcomes. ICC values higher than 0.85 are considered good and measures below 0.7 as poor . To further inspect the differences in power output Bland-Altman plots and limits of agreement were used .
Finally the total mean power output measured from the wheels was compared to that estimated by the drag test - pulley combination using an Anova for the different mean power outputs of the different measurement systems. Overall statistical significance was set at p<0.05.
The nine male subjects had a mean age of 25.9 years (std = 9.6), a mean body mass of 90.3 kg (std = 12.5) and a mean height of 1.90 m (std = 0.04). All subjects enrolled in the study after 8 sessions (1 first session of 12min and 7sessions of 8min) of low-intensity steady-state wheelchair exercise on the motor driven treadmill.
Cross correlation and the corresponding phase lag between the two torque signals around the wheel axis, for the different trials
Means and standard deviation (between brackets) of propulsion characteristics for the different measurement-wheels (Optipush (Op) en Smartwheel (Sw)) over the three 4-minute blocks (n=9 AB subjects)
ICC Op Within
ICC Sw Within
Push time (s)
Cycle time (s)
Contact angle (rad)
Mean Power/push (W)
Mean power/ minute (W)
Bland altman plots
Measurement-wheels and drag test
Power output (Pout) as determined by three different methods. There is a significant difference between the measurement-wheels and the drag test
Mean Pout x2 Op(Watt)
Mean Pout x2 Sw(Watt)
Pout dragtest (Watt)
The aim of the present study was to compare two different measurement-wheels with supposedly the same data output under a real life dynamic, yet standardized submaximal wheeling condition. The results will help interpretation in future research regarding wheelchair propulsion with the use of these wheels, and will help to further investigate the intricate interlimb coupling during this bimanual task. The results showed good agreement between both wheels during steady-state propulsion on a treadmill. Both in time (cross-correlation) and in amplitude (intra-class correlation) a high correlation between the wheels was found (Tables 2 and 3). With regard to the power output both wheels showed comparable and consistent results.
The directly measured torque signals had a high cross-correlation, but over the different 4-minute blocks the time lag between the two signals became larger, from 0.04 s after 4 minutes to 0.16 s after 12 minutes. While the wheels were synchronized at the start a synchronous stop was not possible for the Optipush. The two internal clocks of both devices probably differed, yet in the current setup it was not possible to say in which way since a third source of known reliability was not available. Had a synchronous stop been possible we could have corrected for this phenomenon. Despite the small magnitude such options would be greatly appreciated in the future, especially if these measures are to be combined with other measurement systems like EMG or position registration.
Most push characteristics had an ICC higher than 0.9. The timing parameters push time, cycle time and frequency approach an ICC of 1.00 and so did the contact angle. These results all added to the conclusion that the provided signals by both wheels were highly comparable in the time and space profiles. The force-related parameters peak force, work per push and power had slightly lower ICC’s, but were still sufficiently high given that they measured different limb actions, which force profiles had to be produced individually and might also depend on hand dominance [36, 37]. Since the data were collected within a larger framework of experiments it was decided not to change the sides of the different measurement-wheels, which could have shown differences due to hand dominance. Yet the studies that did look into the interlimb coupling and relationship between dominant and non-dominant hand in wheelchair propulsion did not yet show a clear effect of handedness [16, 21]. Future experiments using both wheels could further investigate this possible confounder of the results.
Bland altmann plots
Left-right power output
Power output is a complex output measure, using different components of a measurement-wheel, in this case torque multiplied by angular velocity averaged over time. Assuming a balanced, well-maintained and good quality wheelchair system on a stable and level treadmill, average power output at the left and right side should be identical over time during steady-state wheelchair propulsion on a motor driven treadmill. Although on group level the average difference in power output indeed was almost zero, individual subjects clearly showed differences between the mean power output left and right (Figure 5). Accordingly the ICC for mean power output was lower than for some of the other propulsion characteristics and remarkably lower than the ICC within the measurement-wheels for the three different trials. This means that as expected mean power output of the three different trials was more consistent within the wheels than between wheels.
The individual differences in power output left and right might have different causes. First, dependent on the weight distribution of the subject in the chair and the position fluctuations of the subject-wheelchair combination on the treadmill belt, rolling friction in both the rear wheels and the front castor wheels might be different left and right. More weight on the left or right front castor wheel will increase friction on that side, leading to a higher necessary power output on that side. Second, the weight distribution by the subject between both wheels also proportionally influences the power output; leaning over to one side makes the power output on that side necessarily larger. Third, the pulley system was positioned in line with the center of the treadmill, however subjects propel an approximately 0.75 m wide wheelchair on a 1.20 m wide treadmill, which allowed for movement toward either side of the treadmill. Propelling the chair more to the side will give a force component from the pulley system orthogonal to the wheels resulting in more power output on the outer side with respect to the pulley system. Also, the belt tension of the treadmill is somewhat different at the sides versus the center of the belt, which may also lead to slightly higher levels of rolling resistance when coasting left or right on the treadmill instead of in the middle.
Finally, the wheels under study might be of influence on the measured power output. The differences in mass and inertia of both wheels could potentially have influenced left-right power output. The suppliers did unfortunately not make inertial properties available, which would be greatly appreciated in the future development. Secondly the two wheels could measure torque and angle differently resulting in different power output. While the first three arguments are assumed to be distributed equally over all subjects, the argument of the measurement-wheels is a systematic difference. Although the other reasons might have masked a difference due to the measurement-wheels, the absence of a systematic difference in power output or any of the other outcomes in our view supports the conclusion that differences between subjects seem to be caused rather by their own propulsion behavior and geometrical characteristics than by the measurement systems used.
The agreement in power output as found in the present study is in line with earlier studies [30, 39]. In different experimental setups these studies also found good agreement on power output for both sides. Important in this respect is also the operationalization of the term power output. As mentioned in the introduction Hurd et al.  averaged the power over only three push cycles while the other studies averaged power over more cycles. Therefore their finding of asymmetry in power output, averaged over just three consecutive pushes, seems to be in line with our finding of high variability in the left-right difference in power output (Figure 4).
Measurement-wheels and dragtest
The wheels measured more power than estimated from the dragtest-pulley combination. In addition to the aforementioned consequence of task variation on power output other factors might have contributed to this difference. First the drag test was performed without speed changes, but at a constant speed. Secondly, additional losses when going to the front and back of the treadmill or left-right are not measured by the drag test, but are measured by the wheels. Thirdly, the friction on the front wheels is also dependent on weight distribution. During the drag test subjects were seated in a uniform position (sitting upright with the trunk; hands on the lap), while during propulsion they were free to move in their wheelchair (e.g. with trunk flexion/extension) possibly leading to more rolling friction and thus a higher power. The measurement-wheels seem to measure power in a more accurate way, because they are sensitive to change of torque and angular velocity, still the drag test is the only external comparison currently possible and is relatively cost effective and easy to use
A good agreement between both measurement-wheels was found in this study. Data from both wheels seem consistent and suitable to be used together in experiments on wheelchair propulsion. Both wheels measure a higher mean total power output compared to the estimation of power output using a drag test. If a standardization of an experiment is done using a drag test this should be taken into account. Variability in the execution of wheelchair propulsion seems an essential part in the motor control of this bimanual task. Further research into the interlimb coupling during this bimanual task might use bilateral measurement-wheels to explain and understand the variability between and within the push cycles of both wheels.
a Three Rivers Holdings, Mesa, AZ, USA.
b MAX Mobility, LLC, Antioch, TN, USA.
c Forcelink b.v, Culemborg The Netherlands.
d Double Performance BV, Gouda, The Netherlands.
Analysis of Variance
The Open Access publication of this paper was funded by the Netherlands Organization for Scientific Research (NWO).
- Akbar M, Balean G, Brunner M, Seyler TM, Bruckner T, Munzinger J, Grieser T, Gerner HJ, Loew M: Prevalence of rotator cuff tear in paraplegic patients compared with controls. J Bone Joint Surg Am 2010, 92: 23-30.View ArticlePubMed
- Alm M, Saraste H, Norrbrink C: Shoulder pain in persons with thoracic spinal cord injury: prevalence and characteristics. J Rehabil Med 2008, 40: 277-283. 10.2340/16501977-0173View ArticlePubMed
- van Drongelen S, de Groot S, Veeger HE, Angenot EL, Dallmeijer AJ, Post MW, van der Woude LH: Upper extremity musculoskeletal pain during and after rehabilitation in wheelchair-using persons with a spinal cord injury. Spinal Cord 2006, 44: 152-159. 10.1038/sj.sc.3101826View ArticlePubMed
- Van Den Berg R, De Groot S, Swart KM, Van Der Woude LH: Physical capacity after 7 weeks of low-intensity wheelchair training. Disabil Rehabil 2010,32(26):2244-2252. 10.3109/09638288.2010.535688View ArticlePubMed
- Richter WM, Kwarciak AM, Guo L, Turner JT: Effects of single-variable biofeedback on wheelchair handrim biomechanics. Arch Phys Med Rehabil 2011, 92: 572-577. 10.1016/j.apmr.2010.11.001View ArticlePubMed
- Morrow MM, Hurd WJ, Kaufman KR, An KN: Shoulder demands in manual wheelchair users across a spectrum of activities. J Electromyogr Kinesiol 2010.
- Vanlandewijck Y, Theisen D, Daly D: Wheelchair propulsion biomechanics: implications for wheelchair sports. Sports Med 2001, 31: 339-367. 10.2165/00007256-200131050-00005View ArticlePubMed
- van der Woude LH, de Groot S, Janssen TW: Manual wheelchairs: research and innovation in rehabilitation, sports, daily life and health. Med Eng Phys 2006, 28: 905-915. 10.1016/j.medengphy.2005.12.001View ArticlePubMed
- Cooper RA: SMARTWheel: from concept to clinical practice. Prosthet Orthot Int 2009, 33: 198-209. 10.1080/03093640903082126PubMed CentralView ArticlePubMed
- Kwarciak AM, Rodriguez RE, Sarkar N, Richter WM, Guo A: Validation of a biofeedback system for wheelchair propulsion training. Rehabilitation Research and Practice 2011, 2011: 590780.PubMed CentralPubMed
- De Groot S, Vegter RJK, Vuijk C, van Dijk FW, van der Woude L: WHEEL-i: the development of a wheelchair propulsion lab for rehabilitation and sports. In Everyday Technology for Independence and Care. 29th edition. Edited by: Gelderblom GJ. Amsterdam: IOS Press; 2011.
- Richter WM, Rodriguez R, Woods KR, Karpinski AP, Axelson PW: Reduced finger and wrist flexor activity during propulsion with a new flexible handrim. Arch Phys Med Rehabil 2006, 87: 1643-1647. 10.1016/j.apmr.2006.09.009View ArticlePubMed
- van Drongelen S, van der Woude LH, Janssen TW, Angenot EL, Chadwick EK, Veeger DH: Glenohumeral contact forces and muscle forces evaluated in wheelchair-related activities of daily living in able-bodied subjects versus subjects with paraplegia and tetraplegia. Arch Phys Med Rehabil 2005, 86: 1434-1440. 10.1016/j.apmr.2005.03.014View ArticlePubMed
- Desroches G, Dumas R, Pradon D, Vaslin P, Lepoutre F-X, ChËze L: Upper limb joint dynamics during manual wheelchair propulsion. Clin Biomech 2010, 25: 299-306. 10.1016/j.clinbiomech.2009.12.011View Article
- Limroongreungrat W, Wang YT, Chang L-s, Geil MD, Johnson JT: An instrumented wheel system for measuring 3-D pushrim kinetics during racing wheelchair propulsion. Res Sports Med: An International Journal 2009, 17: 182-194. 10.1080/15438620903120637View Article
- De Groot S, Veeger DH, Hollander AP, Van der Woude LH: Wheelchair propulsion technique and mechanical efficiency after 3 wk of practice. Med Sci Sports Exerc 2002, 34: 756-766. 10.1097/00005768-200205000-00005View ArticlePubMed
- Kwarciak AM, Sisto SA, Yarossi M, Price R, Komaroff E, Boninger ML: Redefining the manual wheelchair stroke cycle: identification and impact of nonpropulsive pushrim contact. Arch Phys Med Rehabil 2009, 90: 20-26. 10.1016/j.apmr.2008.07.013View ArticlePubMed
- Lenton JP, van der Woude LHV, Fowler NE, Nicholson G, Tolfrey K, Goosey-Tolfrey VL: Hand-rim forces and gross mechanical efficiency at various frequencies of wheelchair propulsion. Int J Sports Med 2013,34(2):158.PubMed
- Stergiou N, Decker LM: Human movement variability, nonlinear dynamics, and pathology: Is there a connection? Hum Mov Sci 2011, 30: 869-888. 10.1016/j.humov.2011.06.002PubMed CentralView ArticlePubMed
- Swinnen SP: Intermanual coordination: from behavioural principles to neural-network interactions. Nat Rev Neurosci 2002, 3: 348-359. 10.1038/nrn807View ArticlePubMed
- Hurd WJ, Morrow MM, Kaufman KR, An KN: Biomechanic evaluation of upper-extremity symmetry during manual wheelchair propulsion over varied terrain. Arch Phys Med Rehabil 2008, 89: 1996-2002. 10.1016/j.apmr.2008.03.020PubMed CentralView ArticlePubMed
- Boninger ML, Souza AL, Cooper RA, Fitzgerald SG, Koontz AM, Fay BT: Propulsion patterns and pushrim biomechanics in manual wheelchair propulsion. Arch Phys Med Rehabil 2002, 83: 718-723. 10.1053/apmr.2002.32455View ArticlePubMed
- Asato KT, Cooper RA, Robertson RN, Ster JF: SMART/sup Wheels/: development and testing of a system for measuring manual wheelchair propulsion dynamics. Biomedical Engineering, IEEE Transactions on 1993, 40: 1320-1324. 10.1109/10.250587View Article
- Cooper RA, Boninger ML, VanSickle DP, Robertson RN, Shimada SD: Uncertainty analysis for wheelchair propulsion dynamics. Rehabilitation Engineering, IEEE Transactions on 1997, 5: 130-139. 10.1109/86.593279View Article
- DiGiovine CP, Cooper RA, DiGiovine MM, Boninger ML, Robertson RN: Frequency analysis of kinematics of racing wheelchair propulsion. IEEE Trans Rehabil Eng 2000, 8: 385-393. 10.1109/86.867880View ArticlePubMed
- Newell K, Corcos D: Variability and motor control. Champaign: Human Kinetics; 1993.
- Lamoth CJ, Daffertshofer A, Huys R, Beek PJ: Steady and transient coordination structures of walking and running. Hum Mov Sci 2009, 28: 371-386. 10.1016/j.humov.2008.10.001View ArticlePubMed
- Sparrow WA, Lay BS, O'Dwyer NJ: Metabolic and attentional energy costs of interlimb coordination. J Mot Behav 2007, 39: 259-275. 10.3200/JMBR.39.4.259-275View ArticlePubMed
- Daffertshofer A, Lamoth CJC, Meijer OG, Beek PJ: PCA in studying coordination and variability: a tutorial. Clin Biomech 2004, 19: 415-428. 10.1016/j.clinbiomech.2004.01.005View Article
- van der Woude LH, Bakker WH, Elkhuizen JW, Veeger HE, Gwinn T: Propulsion technique and anaerobic work capacity in elite wheelchair athletes: cross-sectional analysis. Am J Phys Med Rehabil 1998, 77: 222-234. 10.1097/00002060-199805000-00007View ArticlePubMed
- van der Woude LH, de Groot G, Hollander AP, van Ingen Schenau GJ, Rozendal RH: Wheelchair ergonomics and physiological testing of prototypes. Ergonomics 1986, 29: 1561-1573. 10.1080/00140138608967269View ArticlePubMed
- Veeger D, van der Woude LH, Rozendal RH: The effect of rear wheel camber in manual wheelchair propulsion. J Rehabil Res Dev 1989, 26: 37-46.PubMed
- Shrout PE, Fleiss JL: Intraclass correlations: uses in assessing rater reliability. Psychol Bull 1979, 86: 420-428.View ArticlePubMed
- Safrit M, Wood T: Introduction to measurement in physical education and exercise science. 3rd edition. Maryland Heights: Mosby; 1995.
- Bland JM, Altman DG: Statistical methods for assessing agreement between two methods of clinical measurement. Lancet 1986, 1: 307-310.View ArticlePubMed
- de Poel HJ, Peper CE, Beek PJ: Handedness-related asymmetry in coupling strength in bimanual coordination: furthering theory and evidence. Acta Psychol 2007, 124: 209-237. 10.1016/j.actpsy.2006.03.003View Article
- Sainburg RL: Evidence for a dynamic-dominance hypothesis of handedness. Experimental brain research Experimentelle Hirnforschung Experimentation cerebrale 2002, 142: 241-258. 10.1007/s00221-001-0913-8View ArticlePubMed
- Kotajarvi BR, Sabick MB, An KN, Zhao KD, Kaufman KR, Basford JR: The effect of seat position on wheelchair propulsion biomechanics. J Rehabil Res Dev 2004, 41: 403-414. 10.1682/JRRD.2003.01.0008View ArticlePubMed
- Veeger HE, van der Woude LH, Rozendal RH: A computerized wheelchair ergometer. Results of a comparison study. Scand J Rehabil Med 1992, 24: 17-23.PubMed
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.