Analysis of relative displacement between the HX wearable robotic exoskeleton and the user’s hand
© Cempini et al.; licensee BioMed Central Ltd. 2014
Received: 20 February 2014
Accepted: 10 October 2014
Published: 18 October 2014
Advances in technology are allowing for the production of several viable wearable robotic devices to assist with activities of daily living and with rehabilitation. One of the most pressing limitations to user satisfaction is the lack of consistency in motion between the user and the robotic device. The displacement between the robot and the body segment may not correspond because of differences in skin and tissue compliance, mechanical backlash, and/or incorrect fit.
This report presents the results of an analysis of relative displacement between the user’s hand and a wearable exoskeleton, the HX. HX has been designed to maximize comfort, wearability and user safety, exploiting chains with multiple degrees-of-freedom with a modular architecture. These appealing features may introduce several uncertainties in the kinematic performances, especially when considering the anthropometry, morphology and degree of mobility of the human hand. The small relative displacements between the hand and the exoskeleton were measured with a video-based motion capture system, while the user executed several different grips in different exoskeleton modes.
The analysis furnished quantitative results about the device performance, differentiated among device modules and test conditions. In general, the global relative displacement for the distal part of the device was in the range 0.5–1.5 mm, while within 3 mm (worse but still acceptable) for displacements nearest to the hand dorsum. Conclusions over the HX design principles have been drawn, as well as guidelines for future developments.
KeywordsRehabilitation Orthosys Wearability Motion capture Robotic Exoskeleton
In recent years there has been widespread use of robotic systems as advanced rehabilitation tools, both in clinical  and domestic environments . In particular, attention has been focused on actuated orthoses (namely exoskeletons), thanks to their capability to convey powered assistance – either for patient treatment or user empowerment – at the single-joint level [3–6].
Exoskeletons require a shift in the robotic design paradigm towards ergonomics, comfort and safety: related innovative solutions includes series-elastic and variable-stiffness actuator technologies [7, 8], and novel symbiotic control strategies . From the design point of view, endowing the robotic architecture with passive degrees of freedom (DoFs) is a widely used method to achieve enhanced wearability and compliance towards the user’s anthropometry [10, 11]. A recent example comes from the wearable hand exoskeleton HX [12, 13]: its design embeds enough DoFs to cover the variability of human hand gestures and grasps, and has been strongly refined to minimize weight and encumbrance. HX is a modular device, consisting of separable units for the index, thumb, metacarpus, wrist and forearm segments, and can be unlatched from its actuation system.
Having a multitude of DoFs, modules, and an actuation-detaching stage, HX is extremely usable and flexible [14, 15]. Its main drawback is that mechanical stability may be lessened: the main undesired effect is the presence of significant backlash in the human-robot physical interaction.
The aim of the present paper is therefore to analyze the stability of the physical interaction between the user and HX, critically addressing its design approach. The analysis has been conducted in terms of measured kinematic discrepancy between HX and the wearer’s hand; kinematics were acquired with a video-based motion capture system, during several types of grip and with different operative conditions of the robot. Optoelectronic systems have been widely used to measure hand and finger kinematics in both healthy participants [16–18] and patients , and also while they were wearing active exoskeletons [13, 20]. The feasibility of measuring such small displacements, as the ones in the present study, using a video-based motion capture system has been already demonstrated .
Materials and methods
HX hand exoskeleton
Data acquisition and experimental setup
TH: one marker on the IP joint medial side (D), one on the distal phalanx of TH (E) and one on the thumb tip (F);
IF-d (distal): two markers on the distal phalanx of IF (A, B), on the DIP joint and on the tip, respectively, and one on the user’s fingertip (C);
IF-p (proximal): two markers on the proximal phalanx of IF (L, M) and one on the user’s MCP axis (N);
HD: 3 markers on the dorsal shell of the HD (G, H, I) and one on the dorsal aspect of the hand (J), proximal to the knuckle of the middle finger.
active: the exoskeleton moved the hand of the participant, who was requested not to resist the robot’s movement;
passive: the participant moved their hand while wearing the unlatched exoskeleton.
passive, no-hand-movement: global free motion of the upper arm, without moving fingers, hand or wrist;
passive, hand-full-opening: the participant fully opened their hand, reaching a complete extension of the fingers;
minimum and maximum relative movements are expected in these two conditions, respectively. Five trials were performed for each condition, and for each participant. Six healthy adult volunteers participated on the experiments, with their age, weight and height being 39.5 ± 9.1 years, 69.7 ± 7.5 kg, and 1.73 ± 0.05 m (mean ± SD) respectively. All the participants gave written informed consent approved by the Ethics committee of the IRCCS Don Carlo Gnocchi Foundation.
TH: standard deviation of the relative displacement between markers E and F as projected on the longitudinal axis of the distal phalanx (D−F versor);
IF-d: standard deviation of the relative displacement between markers B and C as projected on the longitudinal axis of the distal phalanx (A−B versor);
IF-p: standard deviation of the normal distance of marker M from the index proximal phalanx (segment L−N);
HD: the quadratic mean of the standard deviations of the 3D coordinates of marker J in the dorsum local reference frame (origin in marker H);
Statistical analysis showed that both conditions and hand segments had a significant effect on HX wearing stability (Figure 5). The HD and IF-p showed the greatest displacement, while the IF-d and the TH were respectively just above and always within the noise band. As for the conditions, no-hand-movement and hand-full-opening showed the smallest and largest values respectively, with the two functional tests in between.
Discussion and conclusions
Relative movements between an exoskeleton and the body segments are an indicator of user discomfort and lower robot effectiveness. The results of the present study showed that the stability of HX on the hand is optimal for the distal extremities of the actuated modules, even during the prospective therapeutic use of the device, where the robot drives the hand in executing grasps (“active” mode).Referring to the TH and IF-d plots in Figure 4, the difference between “active” and “passive” can be explained as a drawback of the tendon-driven actuation, which impedes completely transparent behavior because of friction losses in the cable routing. However, the minimal difference between the various modes for IF-d and TH means that HX grasp performance is good, i.e., the robot is stable in driving the fingertip towards closure.
On the contrary, IF-p showed higher displacements in the “active” mode than in the “passive” mode, though the difference was not significant (overlapping CI-95% intervals). This was also because of the sheaths-cable transmission system that, when actuated, pushes the structure while pulling the tendon, generating the revolving torque that is ultimately transmitted to the finger joints. The IF-p values are higher than the IF and TH ones; however, it is worth noting that the IF-p measure highly amplifies the effects of relative unfitting and local backlash in the first phalanx, because of the long geometric distance LM. Moreover, IF-p is the only measure orthogonal to the direction of the hand bones (Figure 3 – HD as well is related to a vector almost lying parallel to the metacarpus), hence it strongly emphasizes misalignment effects. We can state that IF-p represents the worst point for HX-user relative displacement, while IF-d and TH are effective evaluations of the physical grasping stability.
“Active” IF-p measurement remained confined in the 3 mm range (fairly acceptable; since this measure did not depend upon phalanx-parallel translation, but only on phalanx-normal distance variation, we can link it to an angular difference between the HX first phalanx and the user’s one, i.e. to a discrepancy between the MCP joint flexion angles. From the reference exoskeleton values of the LM distance and the angle (40 mm and 35°), variations of 3 mm in the MN projection correspond to a maximum error of ± 5°.
Another critical segment is the HD module, likely because the hand metacarpus is anatomically complex, and it is difficult to wear this module properly. Figure 6 shows that HD displacements are monotonically related to hand aperture. The main undesired effect was the tilting of the HD shell away from the hand dorsum, which (with reference to Figure 3-right) can be interpreted as a rotation of the GHI plane around the y axis (lifting marker I away from the hand dorsum); this was because of the moment reactions coming from the other modules during extension of the fingers, and resulted in a negative increment of the z coordinate and a positive increment of the x coordinate of the J marker.
In conclusion, we have show the statistically significant reliability of HX, through a quantitative analysis of hand and exoskeleton relative movements, especially concerning the finger modules. On the basis of the presented results, future development of HX will focus on refining the fixtures of the HD module, to stabilize its adherence. In the typical “active” operative mode, the displacements at the distal extremities were similar to those registered without motions, and fairly close to the averaged noise level, suggesting that the device can be effective in functional grasp actuation for weak hands. Future studies will also deal with the improvement of the sheath-cable transmission system, limiting friction losses in joint design.
hand exoskeleton device
degrees of freedom
thumb finger module of HX
index finger module of HX
hand dorsum module of HX
confidence interval at 95%
This work was supported in part by the EU within the WAY Project (Wearable interfaces for hand function recoverY, FP7-ICT-Ch5 G.A. 288551), in part by the Italian Ministry of Economic Development within the AMULOS Project (ADVANCED MULOS, Contract MI01 00319), and in part by the Regione Toscana under the Health Regional Research Program 2009 within the project EARLYREHAB.
- Hogan N, Krebs H, Charnnarong J, Srikrishna P, Sharon A: MIT-MANUS: a workstation for manual therapy and training. I. In Proceedings of the IEEE International Workshop on Robot and Human Communication, ROMAN. Tokyo; 1992:161-165.View ArticleGoogle Scholar
- Helal S: Domotics and robotics for supporting seniors. Gerontechnology 2010,9(2):105.View ArticleGoogle Scholar
- Kazerooni H, Steger R, Huang L: Hybrid Control of the Berkeley Lower Extremity Exoskeleton (BLEEX). Int J Robot Res 2006, 25: 561-573. 10.1177/0278364906065505View ArticleGoogle Scholar
- Greenemeier L: Trouble walking? Try Honda’s new exoskeleton legs. Sci Am web blog, November 10, 2008. http://www.scientificamerican.com/blog/post/trouble-walking-try-hondas-new-exos-2008-11-10/?id=trouble-walking-try-hondas-new-exos-2008-11-10
- Jezernik S, Colombo G, Keller T, Frueh H, Morari M: Robotic orthosis lokomat: a rehabilitation and research tool. Neuromodulation: J Int Neuromodulation Soc 2003,6(2):108-115. 10.1046/j.1525-1403.2003.03017.xView ArticleGoogle Scholar
- Nef T, Guidali M, Riener R: ARMin III – arm therapy exoskeleton with an ergonomic shoulder actuation. Appl Bionics Biomech 2009,6(2):127-142. 10.1080/11762320902840179View ArticleGoogle Scholar
- Cempini M, Giovacchini F, Vitiello N, Cortese M, Moise M, Posteraro F, Carrozza MC: NEUROExos: A powered elbow orthosis for post-stroke early neurorehabilitation. Proc Int Conf IEEE Eng Med Biol Soc EMBC 2013, 2013: 342-345.Google Scholar
- Vanderborght B, Albu-Schaeffer A: Variable impedance actuators: A review. Robot Automation Mag 2013,61(12):1601-1614.Google Scholar
- Marchal-Crespo L, Reinkensmeyer DJ: Review of control strategies for robotic movement training after neurologic injury. J Neuroeng Rehabil 2009, 6: 20. 10.1186/1743-0003-6-20View ArticlePubMedPubMed CentralGoogle Scholar
- Schiele A, van der Helm FCT: Kinematic design to improve ergonomics in human machine interaction. IEEE Trans Neural Syst Rehabil Eng 2006,14(4):456-469.View ArticlePubMedGoogle Scholar
- Cempini M, De Rossi SMM, Lenzi T, Vitiello N, Carrozza MC: Self-alignment mechanisms for assistive wearable robots: a kinetostatic compatibility method. IEEE Trans Robot 2013,29(1):236-250.View ArticleGoogle Scholar
- Cempini M, De Rossi SMM, Lenzi T, Mario C, Giovacchini F, Vitiello N, Carrozza MC: Kinematics and design of a portable and wearable exoskeleton for hand rehabilitation. Proc IEEE Int Conf Rehabil Robot ICORR 2013, 2013: 1-6.Google Scholar
- Cempini M, Cortese M, Vitiello N: A powered finger-thumb wearable hand exoskeleton with self-aligning joint axes. Accepted IEEE/ASME Trans Mechatronics 2014. DOI:10:1109/TMECH.2014.2315528Google Scholar
- Chiri A, Cortese M, Rogério de Almeida Riberio P, Cempini M, Vitiello N, Soekadar SR, Carrozza MC: A telerehabilitation system for hand functional training. Converging Clin Eng Res Neurorehabil Biosyst Biorobot 2013, 1: 1019-1023. 10.1007/978-3-642-34546-3_167View ArticleGoogle Scholar
- Cempini M, Chiri A, Rogério de Almeida Ribeiro P, Soekadar SR, Vitiello N, Carrozza MC, Cortese m: A mechatronic system for robot-mediated hand telerehabilitation. Accepted IEEE/ASME Trans Mechatronics 2014. DOI: 10.1109/TMECH.2014.2353298Google Scholar
- Rash G, Belliappa P: A demonstration of the validity of a 3-D video motion analysis method for measuring finger flexion and extension. J Biomechanics 1999,32(12):1337-1341. 10.1016/S0021-9290(99)00140-2View ArticleGoogle Scholar
- Braido P, Zhang X: Quantitative analysis of finger motion coordination in hand manipulative and gestic acts. Hum Movement Sci 2004,22(6):661-678. 10.1016/j.humov.2003.10.001View ArticleGoogle Scholar
- Carpinella I, Mazzoleni P, Rabuffetti M: Experimental protocol for the kinematic analysis of the hand: definition and repeatability. Gait Posture 2006,23(4):445-454. 10.1016/j.gaitpost.2005.05.001View ArticlePubMedGoogle Scholar
- Carpinella I, Jonsdottir J, Ferrarin M: Multi-finger coordination in healthy subjects and stroke patients: a mathematical modelling approach. J Neuroeng Rehabil 2011, 8: 19. 10.1186/1743-0003-8-19View ArticlePubMedPubMed CentralGoogle Scholar
- Chiri A, Vitiello N, Giovacchini F, Roccella S, Vecchi F, Carrozza MC: Mechatronic design and characterization of the index finger module of a hand exoskeleton for post-stroke rehabilitation. IEEE/ASME Trans Mechatronics 2012,17(5):884-894.View ArticleGoogle Scholar
- Ferrarin M, Stallard J, Palmieri R, Pedotti A: Estimation of deformation in a walking orthosis for paraplegic patients. Clin Biomech 1993,8(5):255-261. 10.1016/0268-0033(93)90035-GView ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.