Development of measurement system for task oriented step tracking using laser range finder
- Tetsuya Matsumura†1,
- Toshiki Moriguchi†2,
- Minoru Yamada3,
- Kazuki Uemura4,
- Shu Nishiguchi3,
- Tomoki Aoyama3 and
- Masaki Takahashi†5Email author
© Matsumura et al.; licensee BioMed Central Ltd. 2013
Received: 14 March 2012
Accepted: 17 May 2013
Published: 22 May 2013
Avoiding a fall requires fast and appropriate step responses, stepping speed as a fall risk indicator has only been assessed in older adults. We have developed a new measurement system that applies a laser range finder to assess temporal and spatial parameters of stepping performance such as step speed, length, and accuracy. This measurement system has higher portability, lower cost, and can analyze a larger number of temporal and spatial parameters than existing measurement systems. The aim of this study was to quantify the system for measuring reaction time and stride duration by compared to that obtained using a force platform.
Ten healthy young adults performed steps in response to visual cues. The measurement system applied a laser range finder to measure the position and velocity of the center of each leg and of both legs.
We applied the developed measurement system to the rhythmic stepping exercise and measured reaction time and stride duration. In addition, the foot-off time and foot-contact time were quantified using the measurement system, and compared to the foot-off time and foot-contact time quantified using a force platform.
We confirmed that the measurement system can detect where a participant stood and measured reaction time and stride duration.
Remarkable consistency was observed in the test-retest reliability of the foot-off time and foot-contact time quantified by the measurement system (p < 0.001). The foot-off time and foot-contact time quantified by the measurement system were highly correlated with the foot-off time and foot-contact time quantified by the force platform (reaction time: r = 0.997, stride duration: r = 0.879; p < 0.001).
The new measurement system provided a valid measure of temporal step parameters in young healthy adults.
The validity of the system to measure reaction time and stride duration was evaluated, and confirmed by applying to the rhythmic stepping exercise.
One-third of community-dwelling individuals, aged over 75 years experience at least one fall a year . Falling occurs in various situations of daily life and generally results from an interaction of multiple and diverse risk factors [2, 3]. Falling is a common problem in the growing elderly population and there is a need for effective and convenient fall-risk assessment tools that can be used in community-based fall prevention programs.
Avoiding a fall requires fast and appropriate step responses. Many previous articles have reported that reaction time can identify elderly individuals at risk of falling [4–8]. The step responses used to avoid a fall involve motor functions, cognitive functions, and visuospatial skills. In previous studies, reaction time, step length, and cognitive function including visuospatial skills and visuospatial working memory affected balance control and were important indicators of fall risk [9–15]. Effective fall risk assessment may therefore require the analysis of multiple parts of stepping performance such as stepping speed, reaction time, step length and step accuracy, rather than only one individual parameter. In this paper, step accuracy means whether the participant moved to the instructed target square.
Stepping performance is typically measured using a force platform, which is a useful tool with which to assess dynamic balance function and has good reliability . However, this conventional measurement system is fixed and has been used for assessing only stepping speed as a fall risk indicator in elderly. In addition, the use of force platforms is inconvenient, expensive, and can measure only a limited number of parameters for stepping performance.
We have developed a new system for measuring step performance. The system uses a laser range finder (LRF) to assess the position and velocity of the center of each leg and of both legs, and uses this information to determine temporal and spatial parameters of the step including step speed, step length, and step accuracy. The LRF is smaller, less expensive, and more portable than force platforms, and the measurement system may therefore be useful for measuring stepping performance in a community-based fall prevention program.
The aim of this study was to investigate the test-retest reliability and validity of temporal and spatial parameters determined by this new measurement system during voluntary stepping in healthy adults.
Students of Kyoto University were recruited as participants for this study. Ten men volunteered, none of who reported present or previous diseases or injuries associated with gait and/or balance impairments. Informed consent was obtained from all participants prior to participation, in accordance with the guidelines approved by the Kyoto University Graduate School of Medicine (approval number E-880) and the Declaration of Human Rights, Helsinki, 1975.
The measurement system is manufactured by Murata Machinery Ltd. and consists of a LRF, a personal computer (PC), and two calibration poles. The scan period of the LRF is 0.1 s. The LRF can measure the distance within the error ±0.06 m in the stepping field. The center of the stepping field is calibrated by using the two poles. Thus, the environment required for the RSE is easily prepared and do not require the accurate installation of the LRF.
The system measured reaction time after the visual cue was displayed before the participant started to step as a cognitive parameter, and stride duration after the visual cue was displayed before the participant stopped to step as a temporal parameter. The system measured the positions and velocities of the center of each leg and of both legs and determined the accuracy with which the participant performed the task, i.e., stepped to the instructed target as spatial parameters. When a force platform is used to measure them, more than one force platform is needed because commonly-used force platforms do not cover the entire stepping field. However, by using the measurement system, we can measure movement parameters simply and inexpensively (the cost of LRF is about 1,000 dollars and the cost of a force platform is about 50,000 dollars). This system may be useful for measuring the stepping performance and can be used in community-based fall prevention programs.
The PC acquires data from the LRF, calculates the leg detection algorithm and displays a visual cue on a monitor.
In the calibration phase, the coordinate system for the stepping field was configured and the parameters required for the leg detection algorithm were identified.
Where, l square is the length of one square. Thus, the environment required for the RSE was easily prepared and did not require the accurate installation of the LRF.
Main loop phase
The main loop phase consisted of three parts: leg detection by LRF data, the judgement of whether the participant moved to the target square and the display of the next visual cue if the participant stood upright in the center.
Leg detection by the LRF data (leg coordinate data and the center-of-leg coordinates) was determined from distance and angle data obtained from the LRF scan, and was used in the leg detection algorithm proposed by Bellotto . The leg detection algorithm in  measures only the position of the center of the legs; however, the leg detection algorithm in the measurement system measures not only the position of the center of the legs but also the positions of both legs. This is expected to improve the accuracy of the step performance measurements.
The leg detection algorithm is shown in Figure 4. We classified the legs into three leg patterns: LA (two Legs Apart), FS (Forward Straddle) and SL (Single Leg). In classifying the legs into the three leg patterns, we define a, b and c, which are, respectively, the leg width, the maximum step length and the width of the two legs together. These were identified in the calibration phase. First, we scanned a two-dimensional surface with the LRF. Next, the vertical edges were extracted from LRF data. Then, the leg pattern was detected by using three leg parameters and the combination of the arrangement of the edges. Finally, we calculated the position and velocity of the center of each leg and of both legs together using LRF data.
The square on which the participant moved to was determined using the position of the center of each leg and of both legs calculated in the leg detection algorithm and the coordinate system fixed for the stepping field. If the legs of the participant were inside the target square, the symbol “Ο” was displayed. On the other hand, if the leg was outside the target square, the symbol “Δ” was displayed, or if both legs were outside the target square, the symbol “×” was displayed.
We evaluated whether the participant moved to the exact target, and displayed the next visual cue when the participant returned and stood upright in the center of the stepping field.
Stepping performance evaluation phase
The stepping trajectory obtained by the leg detection algorithm is shown in Figure 5. We measured foot-off time and foot-contact time from the position and velocity shown in Figure 6. The foot-off time of the leading leg (TFO1) was defined as the point at which the velocity started to increase. The foot-contact time of the leading leg (TFC1) was defined as the point at which the velocity is less than the threshold. In this paper, the threshold is 0.25 m/s. The reaction time of the leading leg (TR1) was calculated from the time that visual cue was displayed (TCUE) and TFO1. Stride duration of the leading leg (TSD1) was calculated from TFO1 and TFC1. Stride length was calculated from the positions at TFO1 and TFC1.
Thus, in the stepping performance evaluation phase, the velocity was calculated from the positional data of the center of each leg and of both legs after RSE ended. TR1, TSD1 and stride length were calculated from the velocity and displayed immediately on the PC.
Test 1: application to the rhythmic stepping exercise
In order to verify the developed measurement system described above, we applied the system to the RSE and measured the stepping trajectories and stepping performance. Ten participants stepped forward, backward, and sideways (a total of 4 steps), and we measured their steps.
Test 2: statistical analysis of the test using a force platform and the measurement system simultaneously
In order to verify the efficacy of TR1 and TSD1 that were obtained by the developed measurement system, TR1 and TSD1were measured by using a force platform and the system simultaneously. The force platform was a portable Kistler 9286 Force Platform. In case of the force platform, TR1 and TSD1 were calculated according to methods of Melzer et al. . Ten participants stepped forward after the visual cue was displayed four times.
The test-retest reliability of TR1 and TSD1 determined by the system was assessed using the intraclass correlation coefficients ICC1, 1 with TR1 and TSD1 calculated using the force platform data. The criterion-related validity was determined using the Spearman’s correlation coefficient between TR1 and TSD1 obtained by the measurement system and TR1 and TSD1 obtained from the force platform. Data were analyzed using the Statistical Package for the Social Sciences (Windows version 19.0).
Test 1: application to the rhythmic stepping exercise
In addition, we confirmed that we can detect which square the participant stands using data of the center of each leg and of both legs. If we use only data of the center of both legs, we can not judge whether the participant moves to the exact target square.
Test 2: statistical analysis of the test using a force platform and the measurement system simultaneously - test-retest reliability
There was high test-retest reliability of the stepping performance obtained by the measurement system (p < 0.001): TR1 (ICC1, 1 = 0.810; 95% confidence interval (CI), 0.485–0.951); TSD1 (ICC1, 1 = 0.831; 95% CI, 0.542–0.957).
Test 2: statistical analysis of the test using a force platform and the measurement system simultaneously - criterion-related validity
TR1 and TSD1 determined using the developed measurement system were highly correlated with TR1 and TSD1 determined using the force platform: TR1 (r = 0.997, p < 0.001); TSD1 (r = 0.879, p < 0.001).
The results indicate that stepping performance was accurately and reliably measured by the developed system. There was a strong relation between reaction time and stride duration determined using the measurement system and the force platform. The magnitude of reliability was nearly equal to intraclass correlation coefficients of the stepping performance reported by Melzer et al. , who reported ICC of 0.91 for foot-off time and 0.76 for foot-contact time measured using a force platform. As a result, the measurement performance of the developed measurement was equivalent to that of a force platform. The measurement system has many advantages in terms of convenience, cost, and the number of parameters that can be obtained.
Compared to other step performance assessment systems, the measurement system offers several important advantages for fall prevention in communities. It is small, relatively in expensive, highly portable, and has minimal setup meaning that stepping tests can be performed quickly even if a large number of participants are targeted. These advantages make the measurement system ideal for measuring stepping in community settings. The system can obtain spatial parameters such as the stepping trajectory that is difficult to obtain by other step performance assessment systems.
Stepping performance measured by the measurement system can be used as assessment parameters [4–8]. Therefore, the system may be useful for assessing fall risk in fall prevention programs in the future. The relation between the stepping performance measured using the developed measurement system and a fall risk in elderly is under survey, and it is expected to be reported in next paper.
Dynamic balance assessments using force platforms have identified factors that are associated with fall risk and have contributed to the development of rehabilitation programs. Force platform measurements indicated that the increase in step execution time that occurs under dual-task conditions was greater in elderly fallers than in elderly non-fallers , and that step training improved the speed of voluntary step initiation . The measurement system evaluated in the present study may enable researchers and therapists to investigate further details of balance control deficits associated with falling and verify the effects of training programs in elderly and patients at clinical sites conveniently.
We tested the validity of a new measurement system that applied a laser range finder and allowed for convenient assessment of stepping performance. The measurement system is advantageous over current systems of measuring stepping performance in terms of convenience, cost, and the number of parameters that can be obtained. In this study, we applied the leg detection algorithm, and determined temporal and spatial step parameters from the position and velocity of the center of each leg and of both legs. The validity of the system to measure reaction time and stride duration was evaluated according to force platform measures, and was confirmed. This measurement system may be helpful for assessing fall risk indicators in elderly in community and clinical settings.
This work was supported in part by JSPS KAKENHI Grant Number 25709015. We would like to thank the students of the Department of Human Health Sciences at Kyoto University for their help with data collection.
- Tinetti ME, Speechley M, Ginter SE: Risk factors for falls among elderly persons living in the community. N Engl J Med 1988, 319: 1701-1707. 10.1056/NEJM198812293192604View ArticlePubMedGoogle Scholar
- Blake AJ, Morgan K, Bendall MJ: Falls by elderly people at home: prevalence and associated factors. Age Ageing 1988, 17: 365-372. 10.1093/ageing/17.6.365View ArticlePubMedGoogle Scholar
- American Geriatrics Society, British Geriatrics Society, American Academy of Orthopaedic Surgeons Panel on Falls Prevention: Guideline for the prevention of falls in older persons. J Am Geriatr Soc 2001, 49: 664-672. 10.1046/j.1532-5415.2001.49115.xView ArticleGoogle Scholar
- St George RJ, Fitzpatrick RC, Rogers MW, Lord SR: Choice stepping response and transfer times: effects of age, fall risk, and secondary tasks. J Gerontol A Biol Sci Med Sci 2007, 62: 537-542. 10.1093/gerona/62.5.537View ArticlePubMedGoogle Scholar
- Melzer I, Kurz I, Shahar D, Levi M, Oddsson L: Application of the voluntary step execution test to identify elderly fallers. Age Ageing 2007, 36: 532-537. 10.1093/ageing/afm068View ArticlePubMedGoogle Scholar
- Pijnappels M, Delbaere K, Sturnieks DL, Lord SR: The association between choice stepping reaction time and falls in older adults–a path analysis model. Age Ageing 2010, 39: 99-104. 10.1093/ageing/afp200View ArticlePubMedGoogle Scholar
- Melzer I, Oddsson LI: The effect of a cognitive task on voluntary step execution in healthy elderly and young individuals. J Am Geriatr Soc 2004, 52: 1255-1262. 10.1111/j.1532-5415.2004.52353.xView ArticlePubMedGoogle Scholar
- Lord SR, Fitzpatrick RC: Choice stepping reaction time: a composite measure of falls risk in older people. J Gerontol A Biol Sci Med Sci 2001, 56: M627-M632. 10.1093/gerona/56.10.M627View ArticlePubMedGoogle Scholar
- Woolley SM, Czaja SJ, Drury CG: An assessment of falls in elderly men and women. J Gerontol A Biol Sci Med Sci 1997, 52: M80-M87.View ArticlePubMedGoogle Scholar
- Lord SR, Clark RD: Simple physiological and clinical tests for the accurate prediction of falling in older people. Gerontology 1996, 42: 199-203. 10.1159/000213793View ArticlePubMedGoogle Scholar
- Hsiao-Wecksler ET, Robinovitch SN: The effect of step length on young and elderly women’s ability to recover balance. Clin Biomech 2007, 22: 574-580. 10.1016/j.clinbiomech.2007.01.013View ArticleGoogle Scholar
- Lindemann U, Lundin-Olsson L, Hauer K, Wengert M, Becker C, Pfeiffer K: Maximum step length as a potential screening tool for falls in non-disabled older adults living in the community. Aging Clin Exp Res 2008, 20: 394-399.View ArticlePubMedGoogle Scholar
- Maylor EA, Allison S, Wing AM: Effects of spatial and nonspatial cognitive activity on postural stability. Br J Psychol 2001, 92: 319-338. 10.1348/000712601162211View ArticleGoogle Scholar
- Sturnieks DL, St George R, Fitzpatrick RC, Lord SR: Effects of spatial and nonspatial memory tasks on choice stepping reaction time in older people. J Gerontol A Biol Sci Med Sci 2008, 63: 1063-1068. 10.1093/gerona/63.10.1063View ArticlePubMedGoogle Scholar
- Liu-Ambrose TY, Ashe MC, Graf P, Beattie BL, Khan KM: Increased risk of falling in older community-dwelling women with mild cognitive impairment. Phys Ther 2008, 88: 1482-1491. 10.2522/ptj.20080117PubMed CentralView ArticlePubMedGoogle Scholar
- Melzer I, Shtilman I, Rosenblatt N, Oddsson LI: Reliability of voluntary step execution behavior under single and dual task conditions. J Neuroeng Rehabil 2007, 4: 16. 10.1186/1743-0003-4-16PubMed CentralView ArticlePubMedGoogle Scholar
- Yamada M, Tanaka B, Nagai K, Aoyama T, Ichihashi N: Rhythmic stepping exercise under cognitive conditions improves fall risk factors in community-dwelling older adults: Preliminary results of a cluster-randomized controlled trial. Aging Ment Health 2011, 15: 647-653. 10.1080/13607863.2010.551341View ArticlePubMedGoogle Scholar
- Bellotto N, Housheng H: Multisensor-Based Human Detection and Tracking for Mobile Service Robots. IEEE Transactions on Systems, Man and Cybernetics, Part B 2009, 39: 167-181.View ArticleGoogle Scholar
- Rogers MW, Johnson ME, Martinez KM, Mille ML, Hedman LD: Step training improves the speed of voluntary step initiation in aging. J Gerontol A Biol Sci Med Sci 2003,58(1):46-51. 10.1093/gerona/58.1.M46View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.