Biomechanical and functional variation in rat sciatic nerve following cuff electrode implantation
© Restaino et al.; licensee BioMed Central Ltd. 2014
Received: 31 December 2013
Accepted: 7 April 2014
Published: 23 April 2014
Nerve cuff electrodes are commonly and successfully used for stimulating peripheral nerves. On the other hand, they occasionally induce functional and morphological changes following chronic implantation, for reasons not always clear. We hypothesize that restriction of nerve mobility due to cuff implantation may alter nerve conduction.
We quantified acute changes in nerve-muscle electrophysiology, using electromyography, and nerve kinematics in anesthetized Sprague Dawley rat sciatic nerves during controlled hindlimb joint movement. We compared electrophysiological and biomechanical response in uncuffed nerves and those secured within a cuff electrode using analysis of variance (ANOVA) and regression analysis.
Tethering resulting from cuff implantation resulted in altered nerve strain and a complex biomechanical environment during joint movement. Coincident with biomechanical changes, electromyography revealed significantly increased variability in the response of conduction latency and amplitude in cuffed, but not free, nerves following joint movement.
Our findings emphasize the importance of the mechanical interface between peripheral nerves and their devices on neurophysiological performance. This work has implications for nerve device design, implantation, and prediction of long-term efficacy.
The peripheral nervous system (PNS) provides a conduit for signal transduction throughout the body. An array of biomedical devices that interact with the PNS have been developed for both recording and stimulation. Currently used devices, both clinically and in a research environment, include epineural and intrafascicular designs. Epineural designs include simple cylindrical cuffs, self-sizing spiral cuff electrodes, and elliptically shaped flattening electrodes [1–4]. These electrodes seek to non-invasively (relatively) elicit and record action potentials from whole nerve bundles. In contrast, intrafascicular designs seek to maximize recording and stimulation selectivity of specific fascicular bundles, though at the expense of increased invasiveness. Intrafascicular designs include single shank longitudinal [5–8] and transverse ,  electrodes as well as transverse electrode arrays [11–14].
Among the many device choices, cylindrical cuffs remain a popular, versatile, and successful technology. They have been used to house multiple electrode types, from simple extraneural contacts ,  to multichannel intrafascicular arrays . Clinically, uses for cuff electrodes include stimulation and recording from the optic nerve for retinal prostheses [15–18], vagus nerve for seizure prevention [19–25], the hypoglossal nerve for sleep apnea and phrenic nerves for respiratory modulation [26–29]. Recent work has also led to advances in cuffs that allow high selectivity of fiber bundles in both stimulation and recording of motor unit potentials for use in functional electrical stimulation (FES) devices. These devices seek to provide ambulatory assistance to patients with spinal cord injuries and as signals to control prosthetics for amputees , , , [30–34]. The simplicity and cylindrical shape of the cuff design also lends itself as a guiding conduit in nerve regeneration studies , .
Despite their advantages and efficacy, some work has shown chronic damage to the nerve and its environment for reasons that have yet to be fully explained. Specifically, several functional and morphological changes have been reported following cuff implantation, including defective nerve conduction, demyelination, axonal degradation, and tissue encapsulation , [35–45]. These may be associated with entrapment following local compression or tethering from the pulling effect of lead migration, which have been observed following clinical implantation of cuff electrodes , , , . Under normal physiological conditions, nerves glide and stretch predictably to accommodate the rotation of spanned joints. Especially in joint regions, segments of nerves approach and, paradoxically, exceed strain thresholds at which conduction is reversibly or irreversibly altered. These thresholds are typically estimated to be 8-15%, though there is high variability across and even within species [47–55]. It then follows that implant-associated entrapment or tethering may perturb this dynamic and carefully controlled biomechanical environment.
Animal use and experimental groups
All procedures were approved by the Institutional Animal Care and Use Committees (IACUC) at the United States Food and Drug Administration and the University of Maryland, College Park. A rat sciatic nerve model was used to test the effects of cuff electrodes on nerve mechanics and function. 14-week old (~400 g), male Sprague-Dawley rats (Taconic, Inc, Rockville, MD) were clustered into two groups, “implant” (n = 19) and “hook” control (n = 6). Implant group nerves were stimulated with a cuff electrode, while the hook group was stimulated with a subminiature hook electrode (Item ID: 501650, Harvard Apparatus). Within these groups, in a “position control” subgroup (n = 5 implant, n = 6 hook), electrophysiological tests were performed in both reversed experimental order (extended to relaxed) and standard order (relaxed to extended).
Nerve cuff electrode
The rats were anesthetized using 2-3% isoflurane, via inhalation, and placed on a heating pad (NECO-819, setting: Medium) to maintain body temperature throughout the procedure. Adequate anesthetic induction was tested throughout surgery via the toe pinch method.
The initial surgical incision was a proximal-distal cut lateral to the femur path. All subsequent incisions were between muscles or heads of muscles (semitendinosus and quadriceps femoris) to preserve tissue and minimize blood loss. The cuff was implanted approximately 0.5 cm proximal to the trifurcation of the sciatic nerve per methods carefully detailed in the literature, with the rat knee in a neutral configuration , . Care was also taken to align the cuff parallel to the long-axis of the nerve during surgery, and the cuff was tightened to create a secure interface with the underlying nerve. Examination of the distal edge of the cuff at 3×-10× magnification confirmed that there was no apparent lateral or anterior nerve bulge indicative of compression. After cycling joints through their range of motion, we confirmed that there was no cuff slippage or translation based on position of distal cuff edge relative to original position on nerve, and that the cuff returned to its original alignment in a neutral configuration. Two scientists (SMR and EA) independently performed implantations, with validation by the other scientist to ensure consistency. All subsequent measurements were made immediately following completion of implantation procedures, with the entire experiment completed within 30 minutes. Following device implantation, all surgical wounds were closed temporarily with a hemostat, allowing the electrode wires to pass through the incision, but better simulating the in vivo constraints of an intact nerve bed during electromyography or joint positioning. After joint positioning, nerves were exposed for digital photography.
Recordings were taken with subdermal needle electrodes (Grass Technologies) implanted in the interosseus muscles of the foot. All recordings were amplified and filtered with a Grass CP511 AC Amplifier (Grass Technologies) set to 100× amplification and 30-3000 Hz bandpass filter. The limb was positioned for measurements as described above. Recordings were analyzed for repeatability among and between stimulation trains. Lack of repeatability could indicate surgical damage or inappropriate cuff positioning, and therefore data sets with inconsistent or non-repeatable action potentials in either joint position were discarded.
All signals were generated or recorded through National Instruments LabVIEW™, and data processing was done in MATLAB™ (MathWorks, Natick, MA).
Statistics were performed using MATLAB (Mathworks, Natick, MA). Data were initially analyzed for outliers using Peirce’s Criterion, a conservative and probability-based method; two points based on amplitudes and one point based on latencies were removed from the cuffed data set (final: n = 14), while none from the control required removal. Means were compared using paired t-tests or ANOVA (as indicated in subsequent text), and variances were compared using F-test. Linear regression slopes were compared using ANCOVA. Tukey’s HSD post-hoc testing was performed to compare individual groups, and accounted for multiple comparisons. Type I error (α) was set to 0.05 for all comparisons.
Comparison of EMG latencies (in miliseconds) for uncuffed (control) and cuffed nerves
To test for coupling between differences in latency and amplitude, we performed linear regression (Figure 7D). Regression analysis produced a weak positive linear relationship between latency and amplitude in cuffed nerves, in both extended and relaxed positions (relaxed: slope = 0.0015; r^2 = 0.65; extended: slope = 0.0014; r^2 = 0.53). On the other hand, control trials showed an opposite negative relationship between latency and amplitude, in both relaxed and extended positions (relaxed: slope = -0.0017; r^2 = 0.94; extended: slope = -0.0011; r^2 = 0.93). ANCOVA analysis confirmed that cuffing, but not joint position, significantly impacted the amplitude-latency relationship (F = 6.68; p < 0.02).
In this work, we performed kinematic analysis and electromyography to probe acute functional influences of nerve cuff implantation. Cuff implantation constrained normal nerve movement, and resulted in a significant increase in variability in both the functional (latency and amplitude) and biomechanical (nerve strain) response; physiological movement of joints spanned by cuffed nerves may especially impact electrode performance and nerve function. We speculate that examination of such acute effects may hold the potential to ultimately predict chronic neuropathy induced by device implantation.
Biomechanical impacts of cuff implantation
An analysis of strain following joint movement shows significantly different behaviors with and without an implant present, implying a significant shift in the mechanical environment (Figures 5, 6). Despite the absence of compression following cuff implantation in a neutral joint position, the local compression and tethering observed at the cuff edges during joint movement may seed the range of effects that has been described in chronic implantation studies, whose symptoms are analogous to entrapment neuropathies . The compressive action of the cuff is likely compounded by, and in return compounds, natural inflammatory actions due to the presence of a foreign body. Reported responses to compressive effects include axonal degeneration, demyelination, fibrosis, and are sometimes associated with a loss in electrophysiological function , [35–45].
It should be noted that despite the possible repercussions in altering the natural biomechanical environment in the short term, several long-term studies show at least some recovery , ,  with remyelination, axonal regeneration, decreased fibrosis , and recovery of function , . Tissue encapsulation of the cuff, a common effect , [35–45], may serve to stabilize the cuff over the life of the implant and lead to growth and partial recovery of damage done to the nerve . Nerve growth has been shown to be positively influenced by strains induced under natural conditions , ; as a result, it is possible that a sustained increase in nerve tension during chronic cuff implantation may promote mechanically induced growth pathways within the axon leading to ameliorative growth .
Electrophysiological impacts of biomechanical changes
Electrophysiological testing showed consistent, stable and expected CMAP shape, but revealed significant differences in action potential conduction with implantation of cuff electrodes compared with controls. Specifically, latency and amplitude measurements showed a significant increase in variability with the cuff implanted. The regression of EMG latency and amplitude given in Figure 7D provides an interesting contrast between cuffed and uncuffed nerves, and some insight into the nature of the interaction between the cuff and the nerve. The negative relationship in control trials is consistent with the expected decrease in latency with increased amplitude. This relationship is likely due to increased conduction velocities in larger diameter fibers, which innervate larger muscle motor units . The positive relationship found between latency and amplitude following cuff implantation is more complex. We speculate that minimal changes in latency, but higher reductions in amplitude (i.e., left side of regression plot in Figure 7D) are consistent with clinical changes following acute nerve compression over a considerable length of the nerve . This may reflect suppression of conduction in more superficial axons, but not in less compressed axons deeper within the nerve. On the opposite end of the correlation (i.e., right side of regression plot in Figure 7B), highly localized impingement could explain increased latency, resulting from local inefficiency in transmission due to focal ischemia or selective loss of large fibers, but maintenance of compound conduction, and thus amplitude, proximal and distal to the site of tethering , .
Implications for device evaluation and design
On one hand, within reason, it is possible to safely exploit the flexibility of the nerve to increase device performance. Recently developed, an elliptically shaped cuff electrode reshapes the circular nerve by flattening it into a more elliptical shape, effectively reducing the minor axis of the nerve and allowing for isolation of specific fiber bundles which have been pushed towards the electrode surface , . On the other hand, our findings indicate that the biomechanical environment of the nerve-device interface may adversely impact device performance and nerve function. We propose that the high variance present in our data is likely due to variations in the mechanical impact of the cuff on the nerve, which is reliant on the relationship with the cuff and the nerve as well as the cuff and the surrounding musculature. Consistency in implant location is a critical aspect of implantations, but adaptations may also be necessary to match variability in nerve and musculature between patients. Studies of clinical implantations of vagal nerve stimulators have shown significant mechanical impact including tensile and compressive loading [67–70] as well as functional changes such as vocal function, voice alteration, dyspnea, coughing and aspiration , , . Some severity in the resulting functional alterations has been alleviated by better matching lead coil diameter to nerve diameter .
Future cuff designs may also reduce mechanical perturbations through utilization of materials more closely suited to the mechanical properties of the nerve tissue. The epidermal microelectronics, developed by Kim et al. , are designed to match the mechanical properties of human skin to allow long term adhesion and comfort to subjects. Similarly, collagen matrix cuffs have been designed as conduits for growth and repair of damaged nerve tissue. These cuffs match the natural mechanical properties of peripheral nerves, but are also designed to slowly incorporate themselves into the epineurium and surrounding tissues. Ideally, after long term implantation, the effect of the cuff would be negligible and only the mechanical presence of the leads would remain, leaving behind a markedly reduced device footprint , . Progress is still required in development of devices able to stimulate without long lead wires, which have been shown to migrate and tug on nerves in clinical applications , , , . This goal may be realized as wireless power transfer technologies become more widely applicable to biomedical applications.
The altered mechanical environment created by the presence of a cuff electrode has significant acute effects on nerve functionality; future studies will directly correlate acute function with chronic neural structure and function. This study provides an important first step in formally examining the reciprocal relationships between nerve mechanics, nerve architecture, and device implantation, towards more effective peripheral nerve device design and evaluation.
This research was supported by grant CBET1042522 from the National Science Foundation and grant W81XWH1010773 from the U.S. Army Medical Research and Materiel Command. We also acknowledge helpful discussions with members of the Neuromuscular Bioengineering Laboratory, Dr. Cristin Welle and other members of the USFDA Biophysics Laboratory.
The mention of commercial products, their source, or their use in connection with material reported herein is not to be construed as either an actual or an implied endorsement of such products by the U.S. Food and Drug Administration.
- Rodriguez FJ, Ceballos D, Schüttler M, Valero A, Valderrama E, Stieglitz T, Navarro X: Polyimide cuff electrodes for peripheral nerve stimulation. J Neurosci Methods 2000, 98: 105-118.View ArticlePubMedGoogle Scholar
- Loeb GE, Peck RA: Cuff electrodes for chronic stimulation and recording of peripheral nerve activity. J Neurosci Methods 1996, 64: 95-103.View ArticlePubMedGoogle Scholar
- Garde K, Keefer E, Botterman B, Galvan P, Romero MI: Early Interfaced Neural Activity from Chronic Amputated Nerves. Front Neuroengineering 2009, 2: 5.View ArticlePubMed CentralGoogle Scholar
- Fugleholm K, Schmalbruch H, Krarup C: Early peripheral nerve regeneration after crushing, sectioning, and freeze studied by implanted electrodes in the cat. J Neurosci 1994, 14: 2659-2673.PubMedGoogle Scholar
- Lawrence SM, Dhillon GS, Horch KW: Fabrication and characteristics of an implantable, polymer-based, intrafascicular electrode. J Neurosci Methods 2003, 131: 9-26.View ArticlePubMedGoogle Scholar
- Lawrence SM, Dhillon GS, Jensen W, Yoshida K, Horch KW: Acute peripheral nerve recording Characteristics of polymer-based longitudinal intrafascicular electrodes. IEEE Trans Neural Syst Rehabil Eng 2004, 12: 345-348.View ArticlePubMedGoogle Scholar
- Micera S, Navarro X, Carpaneto J, Citi L, Tonet O, Rossini PM, Carrozza MC, Hoffmann KP, Vivo M, Yoshida K, Dario P: On the Use of Longitudinal Intrafascicular Peripheral Interfaces for the Control of Cybernetic Hand Prostheses in Amputees. IEEE Trans Neural Syst Rehabil Eng 2008, 16: 453-472.View ArticlePubMedGoogle Scholar
- Navarro X, Lago N, Vivo M, Yoshida K, Koch KP, Poppendieck W, Micera S: Neurobiological evaluation of thin-film longitudinal intrafascicular electrodes as a peripheral nerve interface. Rehabi Robot, ICORR 2007 IEEE 10th Int Conf 2007, 2007: 643-649.View ArticleGoogle Scholar
- Badia J, Boretius T, Pascual-Font A, Udina E, Stieglitz T, Navarro X: Biocompatibility of Chronically Implanted Transverse Intrafascicular Multichannel Electrode (TIME) in the Rat Sciatic Nerve. IEEE Trans Biomed Eng 2011, 58: 2324-2332.View ArticleGoogle Scholar
- Boretius T, Badia J, Pascual-Font A, Schuettler M, Navarro X, Yoshida K, Stieglitz T: A transverse intrafascicular multichannel electrode (TIME) to interface with the peripheral nerve. Biosens Bioelectron 2010, 26: 62-69.View ArticlePubMedGoogle Scholar
- Branner A, Normann RA: A multielectrode array for intrafascicular recording and stimulation in sciatic nerve of cats. Brain Research Bulliten 2000, 51: 293-306.View ArticleGoogle Scholar
- Branner A, Stein RB, Normann RA: Selective stimulation and recording using a slanted multielectrode array. Eng Med Biol 1999 21st Annual Conf the 1999 Annual Fall Meeting of the Biomed Eng Soc] BMES/EMBS Conf 1999, 371: 377.Google Scholar
- Branner A, Stein RB, Normann RA: Selective Stimulation of Cat Sciatic Nerve Using an Array of Varying-Length Microelectrodes. J Neurophysiol 2001, 85: 1585-1594.PubMedGoogle Scholar
- Clark GA, Ledbetter NM, Warren DJ, Harrison RR: Recording sensory and motor information from peripheral nerves with Utah Slanted Electrode Arrays. Eng Med Biol Soc EMBC, 2011 Annual Int Conf the IEEE 2011, 2011: 4641-4644.View ArticleGoogle Scholar
- Delbeke J, Oozeer M, Veraart C: Position, size and luminosity of phosphenes generated by direct optic nerve stimulation. Vision Res 2003, 43: 1091-1102.View ArticlePubMedGoogle Scholar
- Delbeke J, Wanet-Defalque MC, Gérard B, Troosters M, Michaux G, Veraart C: The microsystems based visual prosthesis for optic nerve stimulation. Artif Organs 2012, 26: 232-234.View ArticleGoogle Scholar
- Veraart C, Raftopoulos C, Mortimer JT, Delbeke J, Pins D, Michaux G, Vanlierde A, Parrini S, Wanet-Defalque M-C: Visual sensations produced by optic nerve stimulation using an implanted self-sizing spiral cuff electrode. Brain Res 1998, 813: 181-186.View ArticlePubMedGoogle Scholar
- Veraart C, Wanet-Defalque MC, Gérard B, Vanlierde A, Delbeke J: Pattern Recognition with the Optic Nerve Visual Prosthesis. Artif Organs 2012, 27: 996-1004.View ArticleGoogle Scholar
- DeGiorgio CM, Schachter SC, Handforth A, Salinsky M, Thompson J, Uthman B, Reed R, Collins S, Tecoma E, Morris GL, Vaughn B, Naritoku DK, Henry T, Labar D, Gilmartin R, Labiner D, Osorio I, Ristanovic R, Jones J, Murphy J, Ney G, Wheless J, Lewis P, Heck C: Prospective Long‒Term Study of Vagus Nerve Stimulation for the Treatment of Refractory Seizures. Epilepsia 2012, 41: 1195-1200.View ArticleGoogle Scholar
- Krahl SE, Clark KB, Smith DC, Browning RA: Locus Coeruleus Lesions Suppress the Seizure‒Attenuating Effects of Vagus Nerve Stimulation. Epilepsia 2012, 39: 709-714.View ArticleGoogle Scholar
- Krahl SE, Senanayake SS, Pekary AE, Sattin A: Vagus nerve stimulation (VNS) is effective in a rat model of antidepressant action. J Psychiatr Res 2004, 38: 237-240.View ArticlePubMedGoogle Scholar
- Naritoku DK, Terry WJ, Helfert RH: Regional induction of fos immunoreactivity in the brain by anticonvulsant stimulation of the vagus nerve. Epilepsy Res 1995, 22: 53-62.View ArticlePubMedGoogle Scholar
- Takaya M, Terry WJ, Naritoku DK: Vagus Nerve Stimulation Induces a Sustained Anticonvulsant Effect. Epilepsia 2012, 37: 111-1116.Google Scholar
- Lundgren J, Amark P, Blennow G, Stromblad LG, Wallstedt L: Vagus nerve stimulation in 16 children with refractory epilepsy. Epilepsia 1998, 39: 809-813.View ArticlePubMedGoogle Scholar
- Lundy DS, Casiano RR, Landy HJ, Gallo J, Gallo B, Ramsey RE: Effects of vagal nerve stimulation on laryngeal function. J Voice 1993, 7: 359-364.View ArticlePubMedGoogle Scholar
- Sahin M, Durand DM: Improved nerve cuff electrode recordings with subthreshold anodic currents. IEEE Trans Biomed Eng 1998, 45: 1044-1050.View ArticlePubMedGoogle Scholar
- Sahin M, Haxhiu MA, Durand DM, Dreshaj IA, Sahin M, Haxhiu MA, Durand DM, Dreshaj IA: Spiral nerve cuff electrode for recordings of respiratory output. J Appl Physiol 1997, 83: 317-322.PubMedGoogle Scholar
- DiMarco AF, Onders RP, Kowalski KE, Miller ME, Ferek S, Mortimer JT: Phrenic Nerve Pacing in a Tetraplegic Patient via Intramuscular Diaphragm Electrodes. Am J Respir Crit Care Med 2002, 166: 1604-1606.View ArticlePubMedGoogle Scholar
- Weese-Mayer DE, Silvestri JM, Kenny AS, Ilbawi MN, Hauptman SA, Lipton JW, Talonen PP, Garcia HG, Watt JW, Exner G, Baer GA, Elefteriades JA, Peruzzi WT, Alex CG, Harlid R, Vincken W, Davis GM, Decramer M, Kuenzle C, Saeterhaug A, Schöber JG: Diaphragm Pacing with a Quadripolar Phrenic Nerve Electrode: An International Study. Pacing Clin Electrophysiology 2012, 19: 1311-1319.View ArticleGoogle Scholar
- Gustafson KJ, Pinault GCJ, Neville JJ, Syed I, Davis JA, Jean-Claude J, Triolo RJ: Fascicular anatomy of human femoral nerve: Implications for neural prostheses using nerve cuff electrodes. J Rehabil Res Dev 2009, 46: 973-984.View ArticlePubMedPubMed CentralGoogle Scholar
- Navarro X, Krueger TB, Lago N, Micera S, Stieglitz T, Dario P: A critical review of interfaces with the peripheral nervous system for the control of neuroprostheses and hybrid bionic systems. J Peripher Nerv Syst 2012, 10: 229-258.View ArticleGoogle Scholar
- Frankel MA, Dowden BR, Mathews VJ, Normann RA, Clark GA, Meek SG: Multiple-Input Single-Output Closed-Loop Isometric Force Control Using Asynchronous Intrafascicular Multi-Electrode Stimulation. IEEE Trans Neural Syst Rehabil Eng 2011, 19: 325-332.View ArticlePubMedGoogle Scholar
- Caparso AV, Durand DM, Mansour JM, Caparso AV, Durand DM, Mansour JM: A Nerve Cuff Electrode for Controlled Reshaping of Nerve Geometry. J Biomater Appl 2009, 24: 247-273.View ArticlePubMedPubMed CentralGoogle Scholar
- Polasek KH, Hoyen HA, Keith MW, Tyler DJ: Human Nerve Stimulation Thresholds and Selectivity Using a Multi-contact Nerve Cuff Electrode. IEEE Trans Neural Syst Rehabil 2007, 15: 76-82. EngView ArticleGoogle Scholar
- Edell DJ, Churchill JN, Gourley IM: Biocompatibility of a Silicon Based Peripheral Nerve Electrode. Biomater Med Devices Artif Organs 1982, 10: 103-122.View ArticlePubMedGoogle Scholar
- Murphy B, Krieger C, Hoffer J-A: Chronically implanted epineural electrodes for repeated assessment of nerve conduction velocity and compound action potential amplitude in rodents. J Neurosci Methods 2004, 132: 25-33.View ArticlePubMedGoogle Scholar
- Larsen JO, Thomsen M, Haugland M, Sinkjaer T: Degeneration and Regeneration in rabbit peripheral nerve with long-term nerve cuff electrode implant: a stereological study of myelinated and unmyelinated axons. Acta Neuropathol 1998, 96: 365-378.View ArticlePubMedGoogle Scholar
- Grill WM, Mortimer JT: Electrical Properties of Implant Encapsulation Tissue. Ann Biomed Eng 1994, 22: 23-33.View ArticlePubMedGoogle Scholar
- Leventhal DK, Cohen M, Durand DM: Chronic histological effects of the flat interface nerve electrode. J Neural Eng 2006, 3: 102.View ArticlePubMedGoogle Scholar
- Vince V, Brelen ME, Delbeke J, Colin IM: Anti-TNF-α reduces the inflammatory reaction associated with cuff electrode implantation around the sciatic nerve. J Neuroimmunol 2005, 165: 121-128.View ArticlePubMedGoogle Scholar
- Vince V, Thil M-A, Gérard A-C, Veraart C, Delbeke J, Colin IM: Cuff electrode implantation around the sciaticnerve is associated with an upregulation of TNF-α and TGF-β1. J Neuroimmunol 2005, 159: 75-86.View ArticlePubMedGoogle Scholar
- Thil M-A, Duy DT, Colin IM, Delbeke J: Timecourse of tissue remodelling and electrophysiology in the rat sciatic nerve after spiral cuff electrode implantation. J Neuroimmunol 2007, 185: 103-114.View ArticlePubMedGoogle Scholar
- Grill WM, Mortimer JT: Neural and connective tissue response to long-term implantation of multiple contact nerve cuff electrodes. J Biomed Mater Res 2000, 50: 215-226.View ArticlePubMedGoogle Scholar
- Gupta R, Nassiri N, Hazel A, Bathen M, Mozaffar T: Chronic nerve compression alters Schwann cell myelin architecture in a murine model. Muscle Nerve 2012, 45: 231-241.View ArticlePubMedPubMed CentralGoogle Scholar
- Gupta R, Rowshan K, Chao T, Mozaffar T, Steward O: Chronic nerve compression induces local demyelination and remyelination in a rat model of carpal tunnel syndrome. Exp Neurol 2004, 187: 500-508.View ArticlePubMedGoogle Scholar
- Zumsteg D, Jenny D, Wieser HG: Vocal cord adduction during vagus nerve stimulation for treatment of epilepsy. Neurology 2000, 54: 1388-1389.View ArticlePubMedGoogle Scholar
- Boyd BS, Puttlitz C, Gan J, Topp KS: Strain and excursion in the rat sciatic nerve during a modified straight leg raise are altered after traumatic nerve injury. J Orthop Res 2012, 23: 764-770.View ArticleGoogle Scholar
- Topp KS, Boyd BS: Structure and Biomechanics of Peripheral Nerves: Nerve Responses to Physical Stresses and Implications for Physical Therapist Practice. Phys Ther 2006, 86: 92-109.PubMedGoogle Scholar
- Wright TW, Glowczewskie F Jr, Cowin D, Wheeler DL: Ulnar nerve excursion and strain at the elbow and wrist associated with upper extremity motion. J Hand Surg [Am] 2001, 26: 655-662.View ArticleGoogle Scholar
- Wright TW, Glowczewskie F, Wheeler D, Miller G, Cowin D: Excursion and strain of the median nerve. J Bone Joint Surg Am 1996, 78: 1897-1903.PubMedGoogle Scholar
- Wall EJ, Kwan MK, Rydevik BL, Woo SL, Garfin SR: Stress relaxation of a peripheral nerve. J Hand Surg [Am] 1991, 16: 859-863.View ArticleGoogle Scholar
- Kwan MK, Wall EJ, Massie J, Garfin SR: Strain, stress and stretch of peripheral nerve. Rabbit experiments in vitro and in vivo. Acta Orthop Scand 1992, 63: 267-272.View ArticlePubMedGoogle Scholar
- Phillips JB, Smit X, De Zoysa N, Afoke A, Brown RA: Peripheral nerves in the rat exhibit localized heterogeneity of tensile properties during limb movement. J Physiol 2004, 557: 879-887.View ArticlePubMedPubMed CentralGoogle Scholar
- Wall E, Massie J, Kwan M, Rydevik B, Myers R, Garfin S, Wall E, Massie J, Kwan M, Rydevik B, Myers R, Garfin S: Experimental stretch neuropathy. Changes in nerve conduction under tension. J Bone Joint Surg (Br) 1992, 74: 126-129.Google Scholar
- Highet WB, Sanders FK: The effects of stretching nerves after suture. Br J Surg 2012, 30: 355-369.View ArticleGoogle Scholar
- Naples GG, Mortimer JT, Scheiner A, Sweeney JD: A spiral nerve cuff electrode for peripheral nerve stimulation. IEEE Trans on Biomed Eng 1988, 35: 905-916.View ArticleGoogle Scholar
- Rydevik BL, Kwan MK, Myers RR, Brown RA, Triggs KJ, Woo SL, Garfin SR: An in vitro mechanical and histological study of acute stretching on rabbit tibial nerve. J Orthop Res 1990, 8: 694-701.View ArticlePubMedGoogle Scholar
- Rempel DM, Diao E: Entrapment neuropathies: pathophysiology and pathogenesis. J Electromyogr Kinesiol 2004, 14: 71-75.View ArticlePubMedGoogle Scholar
- Nishiura Y, Hara Y, Yoshii Y, Ochiai N: Gradual stretching of the proximal nerve stump induces the growth of regenerating sprouts in rats. J Orthop Res 2008, 26: 1012-1017.View ArticlePubMedGoogle Scholar
- Nishiura Y, Yamada Y, Hara Y, Ichimura H, Yoshii Y, Ochiai N: Repair of peripheral nerve defect with direct gradual lengthening of the proximal nerve stump in rats. J Orthop Res 2006, 24: 2246-2253.View ArticlePubMedGoogle Scholar
- Loverde JR, Ozoka VC, Aquino R, Lin L, Pfister BJ: Live imaging of axon stretch growth in embryonic and adult neurons. J Neurotrauma 2011, 28: 2389-2403.View ArticlePubMedGoogle Scholar
- Kupa EJ, Roy SH, Kandarian SC, Luca CJD: Effects of muscle fiber type and size on EMG median frequency and conduction velocity. J Appl Physiol 1995, 79: 23-32.PubMedGoogle Scholar
- Stecker MM, Baylor K, Chan YM: Acute nerve compression and the compound muscle action potential. J Brachial Plex and Peripher Nerve Inj 2008, 3: 1.Google Scholar
- Parry GJ, Cornblath DR, Brown MJ: Transient conduction block following acute peripheral nerve ischemia. Muscle Nerve 1985, 8: 409-412.View ArticlePubMedGoogle Scholar
- Kiechl S: 2013. Clinical and Electrodiagnostic Work-up of Peripheral Nerve Lesions. In Medical Radiology, Diagnostic Imaging: High-Resolution Sonography of the Peripheral Nervous System Berlin. Heidelberg: Springer-Verlag Berlin Heidelberg; 2008:43-70.View ArticleGoogle Scholar
- Schiefer MA, Polasek KH, Triolo RJ, Pinault GCJ, Tyler DJ: Selective stimulation of the human femoral nerve with a flat interface nerve electrode. J Neural Eng 2010, 7: 26006.View ArticlePubMedPubMed CentralGoogle Scholar
- Pearl PL, Conry JA, Yaun A, Taylor JL, Heffron AM, Sigman M, Tsuchida TN, Elling NJ, Bruce DA, Gaillard WD: Misidentification of vagus nerve stimulator for intravenous access and other major adverse events. Pediatr Neurol 2008, 38: 248-251.View ArticlePubMedGoogle Scholar
- Ng WH, Donner E, Go C, Abou-Hamden A, Rutka JT: Revision of vagal nerve stimulation (VNS) electrodes: review and report on use of ultra-sharp monopolar tip. Childs Nerv Syst 2010, 26: 1081-1084.View ArticlePubMedGoogle Scholar
- O’Neill BR, Wilberger JE: Revision of vagal nerve stimulator electrodes through a posterior cervical triangle approach: technical note. Neurosurgery 2010, 67: 457-460.PubMedGoogle Scholar
- Ortler M, Unterhofer C, Dobesberger J, Haberlandt E, Trinka E: Complete removal of vagus nerve stimulator generator and electrodes. J Neurosurg Pediatr 2010, 5: 191-194.View ArticlePubMedGoogle Scholar
- Santos PM: Evaluation of laryngeal function after implantation of the vagus nerve stimulation device. Otolaryngol Head Neck Surg 2003, 129: 269-273.View ArticlePubMedGoogle Scholar
- Kim DH, Lu N, Ma R, Kim YS, Kim RH, Wang S, Wu J, Won SM, Tao H, Islam A, Yu KJ, Kim TI, Chowdhury R, Ying M, Xu L, Li M, Chung HJ, Keum H, McCormick M, Liu P, Zhang YW, Omenetto FG, Huang Y, Coleman T, Rogers JA: Epidermal Electronics. Science 2011, 333: 838-843.View ArticlePubMedGoogle Scholar
- Farole A, Jamal BT: A Bioabsorbable Collagen Nerve Cuff (NeuraGen) for Repair of Lingual and Inferior Alveolar Nerve Injuries: A Case Series. J Oral Maxillofac Surg 2008, 66: 2058-2062.View ArticlePubMedGoogle Scholar
- Archibald SJ, Krarup C, Shefner J, Li ST, Madison RD: A collagen‒based nerve guide conduit for peripheral nerve repair: An electrophysiological study of nerve regeneration in rodents and nonhuman primates. J Comp Neurol 2012, 306: 685-696.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.