Personal customizing exercise with a wearable measurement and control unit
© Wang et al; licensee BioMed Central Ltd. 2005
Received: 07 January 2005
Accepted: 28 June 2005
Published: 28 June 2005
Recently, wearable technology has been used in various health-related fields to develop advanced monitoring solutions. However, the monitoring function alone cannot meet all the requirements of customizing machine-based exercise on an individual basis by relying on biosignal-based controls. We propose a new wearable unit design equipped with measurement and control functions to support the customization process.
The wearable unit can measure the heart rate and electromyogram signals during exercise performance and output workload control commands to the exercise machines. The workload is continuously tracked with exercise programs set according to personally customized workload patterns and estimation results from the measured biosignals by a fuzzy control method. Exercise programs are adapted by relying on a computer workstation, which communicates with the wearable unit via wireless connections. A prototype of the wearable unit was tested together with an Internet-based cycle ergometer system to demonstrate that it is possible to customize exercise on an individual basis.
We tested the wearable unit in nine people to assess its suitability to control cycle ergometer exercise. The results confirmed that the unit could successfully control the ergometer workload and continuously support gradual changes in physical activities.
The design of wearable units equipped with measurement and control functions is an important step towards establishing a convenient and continuously supported wellness environment.
In rehabilitation engineering and health promotion, personally customized control of machine-based exercise should be introduced to reflect gradual changes in individual physical work capacity . Biosignal-based workload control systems show great promise as an effective approach to regulate exercise levels [2–4]. Generally, exercise levels are adjusted manually for specific exercise machines, in specific places, typically only by physicians with expertise in sports medicine [5–7]. We have developed an Internet-based cycle ergometer exercise system, aimed at providing a personally customized workload control any time in convenient locations [8, 9]. In this system, exercise resources including exercise programs and workload patterns are distributed over the Internet and dynamically integrated on the cycle ergometer. Workload patterns provided by clinicians are computer files defining the time-course of the exercise to meet individual fitness levels and ability. In practical applications, we prepared and set-up measurement equipment, such as computers, amplifiers, and A/D converters, for individual machines. Unlike these conventional systems, significant advances in wearable technology allow us to continuously assess human biometrics more conveniently. Thus, a wearable unit equipped with measurement and control functions can be used on various machines. That is, by setting up one unit, users can perform biosignal-based exercises at a consistent pace, even on a variety of exercise machines. Accordingly, wearable units have the potential to advance the personal customization process, thereby providing a better exercise routine on an individual basis. A lot of attention has been directed to the investigation of health monitoring services, and various types of wearable unit coordinated monitoring function have been studied [10–14]. Still, there are no wearable units suitable for personally customized machine-based exercise. To implement such units, the workload control function must be embedded into the wearable units, and consequently the units can output control signals to the exercise machines to set the appropriate exercise levels.
Because exercise machines used in gyms/health clubs are configured in very different ways, (e.g., some machines have measurement and control functions, while others do not), most users find it very inconvenient to perform exercise in different places. To provide a personal customizing exercise, we need to measure the biosignals and control the workload without any constraints on machines and locations. Therefore, we separated the measurement and control functions from the exercise machines and incorporated these functions into one wearable unit. This allows the personally customized workload control to be implemented at any convenient place. Another disadvantage of traditional exercise machines is that most of them only provide pre-installed exercise programs with limited variations . This is not cost-efficient because upgrading the exercise programs is very complicated and sometimes impossible. In this case, wearable units equipped with measurement and control functions can be used to loosely couple the exercise machines and programs to easily revise and upgrade conventional exercise programs at end users.
We studied biosignal-based workload control, in which the workload can be adjusted using fuzzy inference to continuously adapt the exercise as a function of heart rate and muscle activity . In this paper, we propose a new design of wearable unit for machined-based exercise. To support the personal customization process, we build the measurement and control functions into a single wearable unit. The unit has several different interfaces for measuring multiple biosignals during exercise and then output control commands to exercise machines. To improve convenience, communications between the exercise machines and the wearable unit are by wireless connections. We developed a prototype of this wearable unit for cycle ergometer exercise and used it as part of an Internet-based exercise system. We examined the wearable unit by recruiting nine volunteers over a two-month period. Our results showed that the wearable unit was effective to handle changes in physical activity while controlling the cycle ergometer and was expected to provide continuously supporting appropriate workload patterns for individuals.
To customize exercise protocols on an individual basis, we need timely updates of workload patterns and continuous workload adjustment, based on the analysis of various biosignals, such as the heart rate (HR) and electromyogram (EMG) signals . Wearable units must offer these measurement and control functions. To enable users to exercise regardless of time and place, the unit must be designed to obtain exercise programs and workload patterns via the Internet and to automatically submit the exercise results.
Wearable Unit Design
Wearable units for machine-based exercise should have interfaces to measure the biosignals. The kind of biosignals required depends on the type of control to be used in exercise programs. We used HR and EMG signals to compute the appropriate exercise levels, according to the idea that gradual changes in physical activity are of interest during an exercise routine. Although exercise programs can be embedded into the wearable unit, they would require a significant amount of the unit's resources, especially if the programs include complicated control methods. Due to the limited processing power and storage capacity available via wearable units, the optimal configuration has wired or wireless communication interfaces to connect to external computers with relatively high performance. If necessary, external computers are utilized for executing exercise programs to provide control parameters. In this case, the wearable unit is a type of middleware, linking the exercise machines to the exercise programs. In addition, like typical designs, the wearable unit needs to have adequate data measurement capacity and transfer speed. Most importantly, the wearable unit should be equipped with an A/D converter and amplifier that operate independently from each exercise machine.
Prototype of a Wearable Unit for Cycle Ergometer Exercise
We developed a prototype of the wearable unit to dynamically control the workload during cycle ergometer exercise. It has a Linux (kernel 2.4) operating system, a 66-MHz-CPU, and 2-MB memory capacity. It also has an on-board 12-bit resolution A/D converter, a 60-dB-gain amplifier, a PCMCIA type slot for a wireless LAN card, and an IP address. Additionally, it features 6 channels for biosignal measurements and a sampling frequency of 5 kHz. At the present development stage, infrared wireless communication is used to acquire HR information from, and output workload control commands to, a cycle ergometer.
Our provided exercise program contains a procedure to calculate the appropriate workload by estimating HR and EMG signals, using a set of predefined fuzzy rules and membership functions . The procedure is time-consuming and requires storage space for the measured data (more than 8-MB for each exercise course). The wearable unit cannot work alone to provide the workload because of its low current capacity. Therefore we used external computers to execute the exercise program and compute the workload. Data transmission between the unit and external computers was implemented using TCP socket communication over wired or wireless connections. At the time of workload control, the unit's built-in low-level control module (Fig. 1) created separate threads to communicate with the external computers and cycle ergometer. Hence, the measurement, control, and data transmission processes were performed individually.
Applying The Wearable Unit to Internet-Based Exercise Systems
We have developed an Internet-based cycle ergometer exercise system [8, 9], which is the backbone of support for the wearable unit, in terms of easy access to various exercise resources at any time from any place. The system provides a central server to process client requests and a history database to store the exercise resources. We have also provided a utility to help clinicians design workload patterns . By coordinating the wearable unit with this system, the practicality and convenience of the personal customization process will improve, because the unit will be able to accommodate various types of cycle ergometers, regardless of whether or not they already have embedded measurement and control functions.
Seven male and two female young subjects (21.3 ± 1.7 years old) assisted us in carrying out the experiments. They exercised once or twice a week for 30 minutes at a time. The exercise flow was the same as for our previous study on the personal customizing exercise . At first, all subjects took a progressively increasing workload test to evaluate their basic physical work capacity. Then, based on the results of this test, a clinician used the design utility to create customized workload patterns by adjusting the fuzzy rules for each subject. The subjects then downloaded the exercise program and the latest workload pattern from the central server and performed the workload control exercise wearing the unit. The workload control exercises by the subjects and the design of the appropriate workload patterns by the clinician were repeatedly performed after the progressively increasing workload test. It should be noted that we provided a web-based user interface to assist the users in obtaining the exercise programs .
Before every exercise session, we downloaded approximately 450-KB of exercise program data as well as 5-KB of workload patterns from the central server to the exercise area. After every session, we uploaded about 8-MB of measured data, including HR and EMG signals, to the central server and stored it in the database.
Wearable Unit for Personally Customized Machine-Based Exercise
Individualized exercise routines are effective for coping with gradual variations in the physical work capacity and for sustaining the motivation to exercise . In machine-based exercise, a practical operation of personal customization is the continuous provision of appropriate workload patterns for users. Thus, when we apply wearable technology to machined-based exercise, the design of the wearable unit must be able to provide the corresponding control function allowing the user to conveniently and easily follow the prescribed workload pattern. However, most wearable unit studies only provide continuous monitoring of various biosignals [10–14], which we believe is insufficient to meet current demands.
We have presented a new wearable unit design equipped with both measurement and control functions for machine-based exercise. The wearable unit gathers measures of the HR and EMG activity and outputs control signals to the exercise machines. Therefore, it is possible to provide appropriate workload control based on individual biosignals. Our results show that a prototype of the wearable unit, combined with an Internet-based exercise system, can achieve personal customization of cycle ergometer exercise. In our experiments, an external computer estimated the appropriate workloads using a biosignal-based fuzzy control method. As a result, the wearable unit formed a link between the user, the exercise machines, and the external computer in which the exercise programs were executed. The wearable unit provided wired and wireless communication interfaces that connected to the external computers. Such designs are very useful if the wearable unit alone cannot perform the computing task in real time. Most importantly, the wearable unit can accommodate various types of cycle ergometers with different specifications, which will greatly improve the convenience of exercising in different places.
The personal customization process has been ensured with the wearable unit. In our experiments, the clinician successfully customized exercise protocols for most of the subjects based on reviewing the subjects' history data. However, two subjects did not perform the anticipated exercises. This had no relationship with the design of the wearable unit, but most likely occurred because our biosignal-based workload control method was not suitable for them. After all, there are great individual differences in terms of functional flexibility and physical work capacity . We require further fundamental studies on providing appropriate exercise levels, based on biosignals. Moreover, cycle ergometer exercise might not be the preferred approach for some subjects. In this case, other types of exercise might be more useful to them.
Information Technology to Support Wearable Units
To continuously support the personally customized workload control without constraints on time and place, the wearable unit must be integrated into an Internet-based support system [1, 9, 19–21], where the exercise routine or design is provided and the measured data is stored and further processed. By transferring the measured data to a central repository, clinicians can review the exercise history and remotely design appropriate workload patterns at their own convenience. Moreover, complicated computing tasks can be assigned to, and the processed results can be acquired from, external computers over wireless connections.
We showed how a wearable unit could be applied to an Internet-based cycle ergometer exercise system. The wearable unit was able to store small amounts of temporal data, and the completed data was processed in an external computer and then uploaded to the database via the Internet. Additionally, the workload patterns and exercise programs were obtained from a central server. Users could perform the individualized exercise routine at any convenient place. Hence, biosignal-based workload control by a wearable unit and the Internet-based support system is a promising approach for providing appropriate exercise levels that will challenge the user and continuously improve their health.
In fact, if we improve the computing performance of the wearable unit by raising the CPU frequency and the internal memory capacity, the unit will be able to compute exercise levels alone. Accordingly, external computers will become unnecessary for control purpose, thus further improving the convenience of the exercise system. For more flexible designs, a removable storage device, which is now being developed, can be used to increase the storage capacity for exercise data and temporal exercise programs. Such design considerations will be implemented in the next version of the wearable unit.
Range of Application in Health Promotion and Rehabilitation
We described how to apply the wearable unit for an indoor cycle ergometer exercise. The wearable unit could also be effective for outdoor exercises, without requiring any significant changes. We investigated the possibility of using biosignals to control power-assisted bicycles . That study attempted to prevent muscular fatigue during cycling by changing the ratio of rider-generated torque to additional electric-motor-produced torque, based on an evaluation of the measured biosignals. The control process approach is similar to cycle ergometer exercise. Thus, by 1) providing an exercise program that implements the control method, and 2) developing control commands to set the assistance ratio, the wearable unit can also be used to support power-assisted bicycle exercise.
Our wearable unit design for machine-based exercise is suitable for health promotion and rehabilitation. The personal customization process provides an ideal approach and facilitates achievement through the increased motivation of the users, who find convenient not to have to worry about whether or not their exercises are suitable. The workload patterns are remotely designed with the help of clinicians, not by self-assessment of users. Moreover, using Internet-based exercise systems with just one unit, users will be able to perform appropriate exercises on exercise machines that have different specifications. The health promotion and rehabilitation industries are expected to receive favorably control-function-equipped wearable units that can dynamically control the exercise levels, based on measured biosignals.
The wearable unit also reduces the costs of developing and producing exercise machines because the measurement and control functions are separate from the machine. Moreover, loosely coupling exercise machines and exercise programs enables the programs can easily be upgraded without tampering with the hardware, i.e., the exercise machines . The wearable unit helps implement such designs in a more flexible manner, because exercise programs can 1) be installed in the wearable unit to directly control the exercise machines, or 2) reside in an external computer used to communicate with the wearable unit to remotely transfer control signals. Moreover, by taking advantage of the wearable unit, the requirements of exercise machines for the personally customized workload control decrease for practical use, and as a result, the possibility of finding a suitable exercise machine without location constraints would increase.
We embedded measurement and control functions into a single wearable unit to personal customizing machine-based exercise. Moreover, we introduced the Internet technology to support the personal customization process without time and place constraints. A wearable unit capable of outputting control signals provides the appropriate exercise levels, based on exercise programs and measured biosignals. Users wearing this unit can take advantage of various exercise programs using a variety of exercise machines. A prototype of the wearable unit measured heart rate and EMG signals and wirelessly transmitted the control commands. By applying this unit to an Internet-based exercise system, we were able to personally customize cycle ergometer exercise. The design of our wearable unit is a progressive step towards establishing a convenient and continuously supported wellness environment. In the future, we will be able to apply these units to outdoor exercises and rehabilitation.
- Kiryu T, Sasaki I, Shibai K, Tanaka K: Providing appropriate exercise levels for the elderly. IEEE Eng Med Biol Mag 2001,20(6):116-124.View ArticlePubMedGoogle Scholar
- Kiryu T, Takahashi K, Ogawa K: Multivariate analysis of muscular fatigue during biycle ergometer exercise. IEEE Trans Biomed Eng 1997,44(8):665-672.View ArticlePubMedGoogle Scholar
- Glass SC, Knowlton RG, Sanjabi PB, Sullivan JJ: Identifying the integrated electromyographic threshold using different muscles during incremental cycling exercise. J Sports Med Phys Fitness 1998,38(1):47-52.PubMedGoogle Scholar
- Mateika J, Duffin J: The ventilation, lactate and electromyographic thresholds during incremental exercise tests in normoxia, hypoxia and hyperoxia. Eur J Applied Physiol 1994, 69: 110-118.View ArticleGoogle Scholar
- Thompson WR, Benardot D, Jonas S: ACSM fitness book. 3rd edition. Champaign: Human Kinetics; 2003.Google Scholar
- Williford HN, Barfield BR, Lazenby RB, Olson MS: A survey of physicians' attitudes and practices related to exercise promotion. Prev Med 1992,21(5):630-636.View ArticlePubMedGoogle Scholar
- McKenna J, Naylor PJ, McDowell N: Barriers to physical activity promotion by general practitioners and practice nurses. Br J Sports Med 1998,32(3):242-247.PubMed CentralView ArticlePubMedGoogle Scholar
- Kiryu T, Yamaguchi K, Tanaka K, Shionoya A: Internet based system for adjusting cycle ergometer workload to moderate exercise. In Proc 21st Annu Int Conf IEEE/EMBS. Atlanta, GA; 1999:615.Google Scholar
- Wang Z, Shibai K, Kiryu T: An Internet-based cycle ergometer by using distributed computing. In Proc 4th Annu IEEE Conf on ITAB. Birmingham, UK; 2003:82-85.Google Scholar
- Jovanov E, Lords AO, Raskovic D, Cox PG, Adhami R, Andrasik F: Stress monitoring using a distributed wireless intelligent sensor system. IEEE Eng Med Biol Mag 2003,22(3):49-55.View ArticlePubMedGoogle Scholar
- Korhonen I, Parkka J, Gils MV: Health monitoring in the home of the future. IEEE Eng Med Biol Mag 2003,22(3):66-73.View ArticlePubMedGoogle Scholar
- Matsushita S, Oba T, Otsuki K, Toji M, Otsuki J, Ogawa K: A wearable sense of balance monitoring system towards daily health care monitoring. In Proc 7th IEEE Int Symp Wearable Computers (ISWC). New York; 2003:176-83.Google Scholar
- Pentland A: Healthwear: Medical technology becomes wearable. IEEE Computer 2004,37(5):42-49.View ArticleGoogle Scholar
- Anliker U, Ward JA, Lukowicz P, Tröster G, Dolveck F, Baer M, Keita F, Schenker E, Catarsi F, Coluccini L, Belardinelli A, Shklarski D, Alon M, Hirt E, Schmid R, Vuskovic M: AMON: A wearable multiparameter medical monitoring and alert system. IEEE Trans Inform Technol Biomed 2004,8(4):415-427.View ArticleGoogle Scholar
- Stamford BA: Choosing and using exercise equipment. Phys Sportsmed 1997,25(1):107-108.View ArticleGoogle Scholar
- Wang Z, Kiryu T: Development of evaluation utilities for the Internet-based wellness cycle ergometer system. In Proc IEEE EMBS Asian-Pacific Conf on Biomed Eng. Keihanna, Japan; 2003. 019265-1.pdfGoogle Scholar
- Wang Z, Kiryu T: Design of a web-based health promotion system and its practical implementation for cycle ergometer exercise. In Proc 26th Annu Int Conf IEEE/EMBS. San Francisco, CA; 2004:3330-3333.Google Scholar
- Borg G, Ljunggren G, Ceci R: The increase of perceived exertion, aches and pain in the legs, heart rate and blood lactate during exercise on a bicycle ergometer. Eur J Appl Physiol 1985,54(4):343-349.View ArticleGoogle Scholar
- Siau K: Health care informatics. IEEE Trans Inform Technol Biomed 2003,7(1):1-7.View ArticleGoogle Scholar
- Ammenwerth E, Gräber S, Herrmann G, Bürkle T, König J: Evaluation of health information systems-problems and challenges. Int J Med Inf 2003,71(2):125-135.View ArticleGoogle Scholar
- Blair SN, Franklin BA, Jakicic JM, Kibler B: New vision for health promotion within sports medicine. Am J Health Promot 2003,18(2):182-185.View ArticlePubMedGoogle Scholar
- Kiryu T, Irishima K, Moriya T, Mizuno Y: Changes in functional activity with prediction during cycling exercise. In Proc Congress of the International Society of Electrophysiology and Kinesiology. Vienna, Austria; 2002:197-198.Google Scholar
- Wang Z, Kiryu T, Iwaki M, Shibai K: An Internet-based cycle ergometer health promotion system for providing personally fitted exercise. IEICE Trans Inf Syst, in press.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.