- Open Access
New method of measuring wrist joint position sense avoiding cutaneous and visual inputs
© Gay et al; licensee BioMed Central Ltd. 2010
Received: 6 May 2009
Accepted: 10 February 2010
Published: 10 February 2010
Aspects of afferent inputs, generally termed proprioception, are being increasingly studied. Extraneous factors such as cutaneous inputs can dramatically interfere while trying to design studies in order to determine the participation of the different structures involved in proprioception in the wrist position sense. We tried to determine validity and repeatability of a new wrist joint position measurement device using methodology designed to minimize extraneous factors and isolate muscle and joint inputs.
In order to test the reliability of the system, eighty young-adult subjects without musculoskeletal or neurologic impairments affecting the right upper extremity were tested using a custom made motion tracking system. Testing consisted of two conditions: active reproduction of active placement and passive reproduction of passive placement. Subjects performed two repetitions of each target position (10, 20, and 30° of flexion and extension) presented in a random order. Test- retest reliability was then tested.
The average constant error in the passive condition was -0.7° ± 4.7° as compared to the active condition at 3.7° ± 5.1°. Average absolute error in the passive condition was 4.9° ± 2.9° compared to the active condition in which absolute error was 5.9° ± 3.5°.
Test-retest repeatability in both conditions was less than the 5° magnitude typical of clinical goniometry. Errors in the active condition (less than 2°) were slightly smaller than the passive condition, and the passive condition was also associated with poorer consistency between apparatus sensors and skin sensors.
The current system for measurement of wrist joint proprioception allows the researcher to decrease extraneous influences that may affect joint position sense awareness, and will help in future study aiming to determine precisely the role of the different structure involved in proprioception.
Aspects of afferent inputs, generally termed proprioception, are being increasingly studied in an attempt to describe and understand impairments , to optimize rehabilitation effectiveness following trauma or surgery , and to prevent recurrent injury [2–9]. Results of previous studies have led to the conclusion that proprioception is multi-faceted and that multiple sensory receptors generate afferent proprioceptive inputs: Visual [10–14], muscle spindle [15–17], cutaneous , tendon and joint . All these receptors have each been demonstrated to contribute to the sense of position or motion of a body part in space [20, 21]. Isolating each proprioceptive input from specific structures in order to determine the effect of disease or injury has proven to be difficult. Clarification of the role and importance of a specific structure such as muscle spindles afferents is essential to understanding the potential impact of surgeries or injuries that diminish or destroy those structures [8, 22, 23].
Methodology differs greatly between studies and even within studies in which a body part is positioned as a target and the same or contralateral body part is positioned to match. Studies also vary in their means of setting the target position, and active, passive, or active-assisted motion may be employed. Regardless of the method used to achieve the target position, the target reproduction may be accomplished by active or passive methods. Theoretically, pairing different types of motions [24–27] could result in a confounding effect. None of these methods of assessing target position reproduction has been adopted as a standard, which likely contributes to variability of results.
Reliability measures for proprioception testing have been in the .85-.95 range at more proximal joints [27–31]. Redundant sensory information may, however, allow the subject to produce more reliable results than may be afforded if extraneous factors are minimized. Techniques utilized to measure joint angles or limb position also present potential confounding factors.
In previous studies, researchers have objectively documented joint position sense using dynamometers [32, 33] electrogoniometers, potentiometers , electromagnetic sensors [6, 21, 35], and video digitization/analysis . Reproducibility of wrist motion measurement using a simple goniometer was reported as 5-8° (intra-observer) and 6-10° (inter-observer) . At the elbow joint, reliability using the electrogoniometer was shown to be superior to either a universal goniometer or a fluid goniometer . The repeatability of electromagnetic sensors is anticipated to be superior to standard goniometric measurements, but has not been demonstrated for the wrist.
The purpose of this study is to formulate a valid and repeatable method for testing wrist joint position sense avoiding stimulating cutaneous inputs. Optimal methodology entails isolating muscle spindles and/or joint receptors contribution to proprioception at the wrist while minimizing extraneous influences using a non-invasive method.
The base of the manipulandum was a Plexiglas disc encasing air jets that, when engaged, allowed frictionless wrist flexion/extension motion over the Plexiglas desktop. A single electromagnetic sensor was attached to the lateral upright of the manipulandum to measure wrist flexion/extension. The wrist joint and sensor alignments were adjusted so that the goniometric and sensor readings all indicated a neutral alignment. All electromagnetic sensors measured position with respect to a source that was mounted anterior and left of the subject on the Plexiglas desktop.
When used for wrist joint position testing, the experimental apparatus was designed to use the sensor located on the manipulandum as an indicator of wrist angle or motion.
Subjects were tested in an "active" and a "passive" condition. Repeatability was calculated by comparing joint excursion measures during two different sessions. Testing positions included two repetitions each of ten, twenty, and thirty degrees of flexion and extension presented in a random order. In the active and passive conditions, the starting position for flexion target angles was wrist extension; and for extension target angles, the starting position was wrist flexion. For all target angles, actual location of the starting position was varied between the positioning and repositioning components of the trial to avoid subjects reproducing the extent of motion rather than joint position. Regardless of the starting position, a minimum excursion of 20° was used for all trials. All the subjects had a training session before starting the experiment in order to minimize the learning effect of the test-retest comparison.
The passive condition began when the examiner gently oscillated the subject's wrist between flexion and extension to assure relaxation. The wrist was moved to a target position and maintained for three seconds while the subject was instructed to remember the position. The wrist was once again oscillated to assure relaxation, passively placed in a different starting position, and then slowly moved toward the target position. The subject was instructed to verbally cue the examiner to stop when the wrist had reached the target position. The target position was then changed and the sequence was repeated. Wrist position was recorded at each stop.
The active condition began by placing the subject's right wrist in a starting position on the opposite side of neutral compared to the target position. The subject was asked to move the hand either "slowly toward your stomach" or "slowly away from your stomach". The subject's motion was stopped when the examiner physically restrained the manipulandum upon reaching the target position and held for three seconds while instructing the subject to remember the position. The subject was instructed not to push against the restraint. The subject was asked to relax and was passively moved to a different starting position. The subject actively returned to the target position at their desired rate of speed, indicating to the examiner when the target position was reached.
Data Collection and Statistical Analysis
First, sample descriptive statistics (means and standard deviation) were calculated for each testing. Signed difference between the targeted position and the repositioned angle (constant error) and the absolute value of that difference (absolute error) were calculated. The standard deviation of the constant error, also known as variable error, was analyzed as an indicator of the consistency of the error. An ANOVA with repeated measure was then used to compare the results in active and passive conditions.
Descriptive statistics of the constant, absolute and variable error for active and passive conditions.
3.7° ± 5.1°
-0.7° ± 4.7°
5.9° ± 3.5°
-4.9° ± 2.9°
Minimizing external influences on proprioceptive input in order to determine the effect of disease or injury has proven to be difficult. These methodological variances make it difficult to reliably isolate and quantify input from a specific structure. Clarification of the role and importance of a specific structure such as muscle spindles afferents is essential to understanding the potential impact of surgeries or injuries that diminish or destroy those structures [8, 22, 23].
Skin-mounted markers or electrogoniometers have been used previously in research and clinical assessment of range of motion [6, 21, 32–36]. These methods were deemed inappropriate for studies attempting to isolate joint contributions to proprioceptive sense because the resultant pressure and cutaneous stretch contribute redundant sensory information about limb position. Our experimental configuration minimized gravitational influences, cutaneous sensory input, and friction to emphasize, if not isolate, muscle spindles contribution to position sense in active versus passive motions. In the current experiment, subjects were not asked to precisely reproduce any joint angles. The data from this study will serve as a baseline measure of experimental setup reliability allowing future studies to differentiate subject variability over and above this demonstrated experimental variability, and to isolate joint or muscle proprioceptive inputs from each other. The technology utilized in this study (3-dimensional Flock of Birds motion tracking system) has a reported accuracy of .5° - 2° [29, 30, 40]. Remaining variability will be attributed to the experimental apparatus and stabilization methodology.
Errors associated with the active condition were similar to the findings of previous studies, which reported errors of 5° or less [25, 27]. Values were superior to range of motion measurements obtained using an instrumented glove designed to capture hand and wrist motions which resulted in repeatability of 6.17° and to goniometry which has been found to be associated with 5-8° intra- and 6-10° inter-rater reliability at the wrist .
Marker placement on the manipulandum appeared to result in more repeatable measures of arcs of motion than did sensors placed directly on the skin. Skin markers may have yielded variable measures because of altered sensor alignment when soft tissues were deformed during motion. For example, during wrist extension, the forearm likely pressed against the lateral support causing the skin to indent and sensor alignment to change. Also, sensor placement on the dorsal forearm could have resulted in slight sensor motion as a result of the motion or stretch on the wrist muscle tendons running under the sensor.
While it was the intent to establish concurrent validity with a tool known to provide accurate measures of motion, we propose that skin electrode placement was not the appropriate tool. It is our contention that since the use of joint repositioning to test proprioception involves comparing the difference between the angle at which the body part is placed and the angle at which it is repositioned, reliable measures are as important as precise measurement of an exact angle. Regardless of sensor placement, the passive condition was consistently associated with less precise repeatability. This methodological variability between active and passive measures forms a baseline of inaccuracy when passive limb placement is paired with active repositioning. This does not take into account the errors anticipated due to different sensory input contributing to joint position sense in active versus passive motion .
One explanation for the methodological differences in reproducibility associated with active and passive motions is that slight extraneous motion may have occurred at the metacarpophalangeal and interphalangeal joints during passive movement of the wrist joint via the manipulandum. In addition, the tubular dressing placed on the fingers could have stretched slightly, allowing a small amount of manipulandum rotation within the palm during passive motion. Another explanation is that the active motion, by stimulating the gamma loop, allows a more precise message encoding by the antagonist muscle spindles Ia fibers [17, 42, 43].
The difference in repeatability between the active and passive conditions lends support for pairing active positioning with active repositioning and passive positioning with passive repositioning when testing joint position sense. Repeatability errors of 1° in the active condition and 3° in the passive condition using manipulandum mounted markers are within acceptable ranges to allow assessment of clinically significant differences of joint position sense.
The importance of proprioception in rehabilitation following musculoskeletal trauma and surgery is becoming increasingly evident, which has lead to a correspondingly increased need to understand the underlying neural mechanisms related to joint mechanics. The system utilized in the current study appears to produce an accurate and repeatable measure of active and passive motion. Differences in variability in active and passive conditions are slight with the current methodology. However, poorer reliability in passive measurements in the skin-mounted sensors lends support for the concept that active and passive motions yield different results. The primary advantage of the current system for measurement of wrist joint proprioception is that it allows the researcher to decrease extraneous influences that may affect joint position sense awareness and therefore improve the knowledge of the mechanisms underlying kinesthesia and proprioception. The results of this study indicate that the measures are repeatable and appear to be equally or more accurate than other measures previously employed to measure wrist and hand range of motion. Nevertheless, other study in order to verify the external validity of this method will be needed.
The authors would like to thanks Kari Hammel for her help in the submission process of the present manuscript.
This study was supported by NIH grant R01 AR047806-02.
Human Kinetics, 1607 N Market St, Champaign, IL 61825.
- Gay A, Parratte S, Salazard B, Guinard D, Pham T, Legre R, Roll JP: Proprioceptive feedback enhancement induced by vibratory stimulation in complex regional pain syndrome type I: An open comparative pilot study in 11 patients. Joint Bone Spine 2007.Google Scholar
- Khudados E, Cody F, O'Boyle D: Proprioceptive regulation of voluntary ankle movements, demonstrated using muscle vibration, is impaired by Parkinson's disease. Journal of Neurology, Neurosurgery, & Psychiatry 1999, 67: 504-510.View ArticleGoogle Scholar
- Rome S, Grünewald R: Abnormal perception of vibration-induced illusion of movement in dystonia. Neurology 1999, 53: 1794-1800.View ArticlePubMedGoogle Scholar
- Konradsen L, Magnusson P: Increased inversion angle replication error in functional ankle instability. Knee Surg Sports Traumatol Arthrosc 2000, 8: 246-251. 10.1007/s001670000124View ArticlePubMedGoogle Scholar
- Miall R, Ingram H, Cole J, Gauthier G: Weight estimation in a "deafferented" man and in control subjects: Are judgments influenced by peripheral or central signals. Exp Brain Res 2000, 133: 491-500. 10.1007/s002210000433View ArticlePubMedGoogle Scholar
- Newcomer K, Laskowski E, Yu B, Johnson J, An K-N: Differences in repositioning error among patients with low back pain compared with control subjects. Spine 2000, 25: 2488-2493. 10.1097/00007632-200010010-00011View ArticlePubMedGoogle Scholar
- Ng G, Chan H: The immediate effects of tension of counterforce forearm brace on neuromuscular performance of wrist extensor muscles in subjects with lateral humeral epicondylitis. Journal of Orthopaedic & Sports Physical Therapy 2004, 34: 72-78.View ArticleGoogle Scholar
- Myers J, Lephart S: Sensorimotor deficits contributing to glenohumreal instability. Clinical Orthopaedics and Related Research 2002, 400: 98-104. 10.1097/00003086-200207000-00013View ArticlePubMedGoogle Scholar
- Willems T, Witvrouw E, Verstuyft J, Vaes P, De Clercq D: Proprioception and muscle strength in subjects with a history of ankle sprains and chronic instability. Journal of Athletic Training 2002, 37: 487-493.PubMed CentralPubMedGoogle Scholar
- Plooy A, Tresilian J, Mon-Williams M, Wann J: The contribution of vision and proprioception to judgements of finger proximity. Exp Brain Res 1998, 118: 415-420. 10.1007/s002210050295View ArticlePubMedGoogle Scholar
- Graziano M: Where is my arm? The relative role of vision and proprioception in the neuronal representation of limb position. Proc Natl Acad Sci USA 1999, 96: 10418-10421. 10.1073/pnas.96.18.10418PubMed CentralView ArticlePubMedGoogle Scholar
- van Beers R, Sittig A, Gon J: Localization of a seen finger is based exclusively on proprioceptoin and on vision of the finger. Exp Brain Res 1999, 125: 43-49. 10.1007/s002210050656View ArticlePubMedGoogle Scholar
- Brindle T, Uhl T, Nitz A, Shapiro R: Motor control of shoulder proprioception at various speeds. Physical Therapy 2003, 33: 109-117.Google Scholar
- Sober S, Sabes P: Multisensory integration during motor planning. The Journal of Neuroscience 2003, 23: 6982-6992.PubMedGoogle Scholar
- Gandevia S, McCloskey D: Joint sense, muscle sense, and their combination as position sense measured at the distal interphalangeal joint of the middle finger. J Phys 1976, 260: 387-407.Google Scholar
- Gandevia S, McCloskey D, Burke D: Kinaesthetic signals and muscle contraction. Trends Neurosci 1992, 15: 62-65. 10.1016/0166-2236(92)90028-7View ArticlePubMedGoogle Scholar
- Roll JP, Vedel JP: Kinaesthetic role of muscle afferents in man, studied by tendon vibration and microneurography. Exp Brain Res 1982, 47: 177-190. 10.1007/BF00239377View ArticlePubMedGoogle Scholar
- Voight ML, Hardin JA, Blackburn TA, Tippett S, Canner GC: The effects of muscle fatigue on and the relationship of arm dominance to shoulder proprioception. J Orthop Sports Phys Ther 1996, 23: 348-352.View ArticlePubMedGoogle Scholar
- Gandevia SC, Burke D: Does the nervous system depend on kinesthetic information to control natural limb movements? Behav Brain Sci 1992, 15: 614-632.View ArticleGoogle Scholar
- Verschueren S, Cordo P, Swinnen S: Representation of wrist joint kinematics by the ensemble of muscle spindles from synergistic muscles. J Neurophysiol 1998, 79: 2265-2276.PubMedGoogle Scholar
- Lönn J, Crenshaw AG, Djupsjöbacka M, Pedersen J, Johansson H: Position sense testing: Influence of starting position and type of displacement. Arch Phys Med Rehabil 2000, 81: 592-597. 10.1016/S0003-9993(00)90040-6View ArticlePubMedGoogle Scholar
- Hogervorst T, Brand R: Mechanoreceptors in joint function. J Bone Joint Surg Am 1998, 80-A: 1365-1378.Google Scholar
- Ishii Y, Tojo T, Terajima K, Terashima S, Bechtold JE: Intracapsular components do not change hip proprioception. J Bone Joint Surg Br 1999, 81-B: 345-348. 10.1302/0301-620X.81B2.9104View ArticleGoogle Scholar
- Voigt M, Jakobsen J, Sinkjaer T: Non-noxious stimulation of the glenohumeral joint capsule elicits strong inhibition of active shoulder muscles in conscious human subjects. Neuroscience Letters 1998, 2: 105-108. 10.1016/S0304-3940(98)00665-XView ArticleGoogle Scholar
- Stillman B, McMeeken J, Macdonnell R: After effects of resisted muscle contractoins on the accuracy of joint positoin sense in elite male athletes. Arch Phys Med Rehabil 1998, 79: 1250-1254. 10.1016/S0003-9993(98)90270-2View ArticlePubMedGoogle Scholar
- Ramsay J, Riddoch M: Position-matching in hte upper limb: Professional ballet dancers perform with outstanding accuracy. Clinical Rehabilitation 2001, 15: 324-330. 10.1191/026921501666288152View ArticlePubMedGoogle Scholar
- Dover G, Powers ME: Reliability of joint position sense and force-reproduction measures during internal and external rotation of the shoulder. Journal of Athletic Training 2003, 38: 304-310.PubMed CentralPubMedGoogle Scholar
- Petrella R, Lattanzio P, Nelson M: Effect of age and activity on knee joint proprioception. American Journal of Physical Medicine & Rehabilitation 1997, 76: 235-241.View ArticleGoogle Scholar
- Meskers C, Fraterman H, Helm F, Vermeulen H, Rozing P: Calibration of the "Flock of Birds" electromagnetic tracking device and its application in shoulder motion studies. Journal of Biomechanics 1999, 32: 629-633. 10.1016/S0021-9290(99)00011-1View ArticlePubMedGoogle Scholar
- Ying N, Kim W: Use of dual Euler angles to quantify the three-dimensional joint motion and its application to the ankle joint complex. Journal of Biomechanics 2002, 35: 1647-1657. 10.1016/S0021-9290(02)00241-5View ArticlePubMedGoogle Scholar
- Deshpande N, Connelly D, Culham E, Costigan P: Reliability and validity of ankle proprioceptive measures. Arch Phys Med Rehabil 2003, 84: 883-889. 10.1016/S0003-9993(03)00016-9View ArticlePubMedGoogle Scholar
- Iwasa J, Ochi M, Adachi N, Tobita M, Katsube K, Uchio Y: Proprioceptive improvement in knees with anterior cruciate ligament reconstruction. Clinical Orthopaedics and Related Research 2000, 381: 168-176. 10.1097/00003086-200012000-00020View ArticlePubMedGoogle Scholar
- Janwantanakul P, Magarey M, Jones M, Dansie B: Variation in shoulder position sense at mid and extreme range of motion. Arch Phys Med Rehabil 2001, 82: 840-844. 10.1053/apmr.2001.21865View ArticlePubMedGoogle Scholar
- Cash R, Gonzalez M, Garst J, Barmada R, Stern S: Proprioception after arthroplasty: role of the posterior cruciate. ligament. Clinical Orthopaedics and Related Research 1996, 331: 172-178. 10.1097/00003086-199610000-00024View ArticlePubMedGoogle Scholar
- Newcomer K, Laskowski E, Yu B, Johnson J, An K-N: The effects of a lumbar support on repositioning error in subjects with low back pain. Arch Phys Med Rehabil 2001, 82: 906-910. 10.1053/apmr.2001.23281View ArticlePubMedGoogle Scholar
- Baker V, Bennell K, Stillman B, Cowan S, Crossley K: Abnormal knee joint position sense in individuals with patellofemoral pain syndrome. Journal of Orthopedic Research 2001, 20: 208-214. 10.1016/S0736-0266(01)00106-1View ArticleGoogle Scholar
- Solgaard S, Carlsen A, Kramhoft M, Petersen VS: Reproducibility of goniometry of the wrist. Scand J Rehabil Med 1986, 18: 5-7.PubMedGoogle Scholar
- Goodwin J, Clark C, Deakes J, Burdon D, Lawrence C: Clinical methods of goniometry: a comparative study. Disabil Rehabil 1992, 14: 10-15. 10.3109/09638289209166420View ArticlePubMedGoogle Scholar
- Bland J, Altman D: Statistical methods for assessing agreement between two methods of clinical measurement. Lancet 1986, 307-310.Google Scholar
- Bull AM, Amis AA: Accuracy of an electromagnetic tracking device. Journal of Biomechanics 1997, 30: 857-859. 10.1016/S0021-9290(97)00020-1View ArticlePubMedGoogle Scholar
- Dipietro L, Sabatini A, Dario P: Evaluation of an instrumented glove for hand-movement acquisition. Journal of Rehabilitation Research and Development 2003, 40: 179-190. 10.1682/JRRD.2003.03.0181View ArticlePubMedGoogle Scholar
- Ribot-Ciscar E, Bergenheim M, Albert F, Roll J-P: Proprioceptive population coding of limb position in humans. Exp Brain Res 2003, 149: 512-519.PubMedGoogle Scholar
- Roll JP, Vedel JP, Roll R: Eye, head and skeletal muscle spindle feedback in the elaboration of body references. Prog Brain Res 1989, 80: 113-123.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.