- Short report
- Open Access
Wearing a safety harness during treadmill walking influences lower extremity kinematics mainly through changes in ankle regularity and local stability
Journal of NeuroEngineering and Rehabilitation volume 9, Article number: 8 (2012)
Wearing a harness during treadmill walking ensures the subject's safety and is common practice in biomedical engineering research. However, the extent to which such practice influences gait is unknown. This study investigated harness-related changes in gait patterns, as evaluated from lower extremity kinematics during treadmill walking.
Healthy subjects (n = 10) walked on a treadmill at their preferred speed for 3 minutes with and without wearing a harness (LiteGait®, Mobility Research, Inc.). In the former condition, no weight support was provided to the subjects. Lower extremity kinematics was assessed in the sagittal plane from the mean (meanRoM), standard deviation (SDRoM) and coefficient of variation (CoVRoM) of the hip, knee, and ankle ranges of motion (RoM), as well as from the sample entropy (SampEn) and the largest Lyapunov exponent (LyE) of the joints' angles. Wearing the harness increased the meanRoM of the hip, the SDRoM and the CoVRoM of the knee, and the SampEn and the LyE of the ankle. In particular, the harness effect sizes for both the SampEn and the LyE of the ankle were large, likely reflecting a meaningful decline in the neuromuscular stabilizing control of this joint.
Wearing a harness during treadmill walking marginally influences lower extremity kinematics, resulting in more or less subtle changes in certain kinematic variables. However, in cases where differences in gait patterns would be expressed through modifications in these variables, having subjects walk with a harness may mask or reinforce such differences.
Treadmill walking is commonly used for biomedical engineering research and rehabilitation purposes. In research, it allows investigators to acquire gait variables from a large number of consecutive steps in an easy and time-saving manner . In rehabilitation, when combined with a body weight support (BWS) system, it enables patients who are unable to fully bear their weight to safely initiate a retraining program . Investigations have thus examined whether gait patterns are equivalent between treadmill and overground walking [3, 4], and kept normal while walking with a BWS system [5, 6]. Overall, treadmill and overground walking have been shown to be mechanically similar , but with reduced variability and improved local stability of the lower extremities during treadmill walking [4, 5]. On the other hand, increasing the level of BWS has been found to progressively alter gait kinematics and kinetics [6, 7], with more pronounced changes in the kinetic patterns .
However, an issue related to treadmill walking that has been poorly investigated concerns the effect of the harness alone (0% BWS) on gait. Ivanenko et al.  indicated that the kinematic and muscle activity patterns of the lower extremities during treadmill walking with a harness at 0% BWS were roughly similar to previous findings they obtained using a setup where subjects walked on a treadmill without harness . Although these findings provide indirect evidence that the impact of the harness alone on gait patterns is likely limited, no definite conclusion can be drawn since the effect of wearing a harness was not examined per se. Recently, Aaslund and Moe-Nilssen  specifically investigated the effect of wearing a harness on trunk movements during gait and demonstrated a reduction in the vertical trunk acceleration. Since damping of vertical trunk acceleration is predominantly achieved by the lower extremities, the authors suggested that significant modulations of lower extremity control may occur when wearing a harness. Therefore, our study aimed to explore this issue. We hypothesized that wearing a harness would be associated with significant changes in gait patterns, as evaluated from lower extremity kinematics.
Ten healthy right leg dominant subjects (4 females and 6 males; age: 24.8 ± 4.0 years; body weight: 80.5 ± 18.41 kg; height: 1.78 ± 0.10 m) participated in the study after signing an institutionally approved informed consent form. The subjects were free of lower extremity injuries and disabilities that might influence walking ability. Reflective markers were attached to a tight fitting suit at specific anatomical landmarks on the lower extremities [10, 11] (Figure 1). After having determined the subjects' preferred walking speed (PWS; mean ± SD of the group: 1.11 ± 0.16 m.s-1) on treadmill (312-C, Bodyguard) using a well-established protocol , two treadmill walking conditions, presented randomly, were performed at PWS: wearing and without wearing a harness. For the harness condition, participants were fitted into the LiteGait® partial weight bearing system (Mobility Research, Inc.) (Figure 1). The system was adjusted to not support the subject (0% BWS), as measured and monitored using the BiSym, a digital microprocessor that displays in real-time the load support provided by the LiteGait®. For each condition, subjects walked for 3 minutes.
The three-dimensional marker positions were acquired (60 Hz) with an 8-camera Motion Analysis Eagle Digital system. The anatomical joint angles of the right/left hips, knees and ankles were then obtained in the sagittal plane using published algorithms . Only this plane of motion was considered since data from the other planes collected via skin markers are associated with increased measurement error . The ranges of motion (RoM) were then identified, from the 3 minute time series, by subtracting the minimum joint angle from the maximum joint angle for each gait cycle. A gait cycle corresponded to the interval between consecutive ipsilateral toe-off events, with the toe-off defined from the maximum backward displacements of the marker located between the second and third metatarsophalangeal joints. For consistency across subjects, the RoM time series were shortened to 134 data points, which was the number of gait cycles of the slowest subject.
Gait function was examined from the joint RoM time series by calculating the mean (meanRoM), the standard deviation (SDRoM) and the coefficient of variation (CoVRoM). From the time series of joint angles were also obtained the largest Lyapunov exponent (LyE) and the sample entropy (SampEn), quantifying local stability and regularity of the joint kinematic patterns, respectively. Smaller LyE and SampEn values reflect more stable and periodic patterns, while larger values reflect more unstable and irregular behaviours, respectively. Details on the algorithms and input parameters used to calculate these measures are provided in Figure 2 and available elsewhere [14–16]. For each joint, the above measures were analyzed using two-way within-subjects ANOVAs (Side: Right/Left; Harness: With/Without). The factor side was included into the analyses to evaluate possible changes in the (a)symmetrical behaviour of the lower extremities due to the harness (a possible side×harness interaction effect). Effect sizes are reported as η2 = SSexplained/SStotal.
Only three out of fifteen ANOVAs revealed a significant harness main effect and two out of fifteen a side×harness interaction effect. For meanRoM, a significant harness effect (F[1, 9] = 18.17; p = 0.002; η2 = 0.18) was observed at the hip, with the harness producing a larger value (36.27 ± 1.15° vs. 35.26 ± 1.02°, Figure 3). For SDRoM and CoVRoM, significant interaction effects (F[1, 9] = 6.12; p = 0.035; η2 = 0.06; and F[1, 9] = 6.37; p = 0.032, η2 = 0.06, respectively) were observed at the knee, with the harness causing larger values at the right knee only (SDRoM: 1.46 ± 0.17° vs. 1.26 ± 0.12°, CoVRoM: 2.51 ± 0.33% vs. 2.16 ± 0.24%, Figure 3). Besides, with harness, the right knee exhibited larger CoVRoM values than its left counterpart (2.51 ± 0.33% vs. 2.26 ± 0.25%, Figure 3).
Finally, a significant harness effect was found for both the SampEn and the LyE of the ankle (F[1, 9] = 7.59; p = 0.022; η2 = 0.14; and F[1, 9] = 9.99; p = 0.011, η2 = 0.31, respectively), with larger values with harness (SampEn: 0.29 ± 0.02 vs. 0.27 ± 0.01; LyE: 1.13 ± 0.09 vs. 0.99 ± 0.07; Figure 4).
Overall, the impact of the harness on lower extremity kinematics has been found to be limited, with most of the measures being similar between the two conditions (with and without harness). However, more or less subtle differences in kinematics have also been observed and deserve to be discussed. First, the increased hip RoM (~1°) with harness indicates a small deviation from normal gait. It also confirms that the LiteGait® system was properly adjusted and did not support the subjects since any BWS leads to decreased joint RoM . Increased hip RoM in the sagittal plane has been observed when carrying external loads [17, 18], which compensates for the decreased pelvic rotation so that walking speed is kept constant . Although the extra load from the harness is negligible in our experiment, the contact of the harness to the waist may have reduced transverse pelvic rotation that was compensated by increasing hip RoM. This assumption calls for examining the effects of the harness on pelvic rotation that together with thoracic rotation and arm movements critically determine trunk momentum and gait stability [19, 20]. However, it is important to not over-enhance this result since a one-degree increase in hip RoM is limited, especially when balanced with measurement errors of kinematic data and intra-subject variability in gait performance. Second, the larger variability observed for the right knee motion with the harness, through increased SDRoM (~0.2°) and CoVRoM (~0.4%), also supports the hypothesis and can be interpreted as a decline in the functional ability to walk [4, 21, 22]. The fact that differences were only observed at the right knee may originate from gait asymmetry [23, 24], and specifically, from functional differences between the lower extremities; the dominant and non-dominant (here, right and left, respectively) lower extremities being mainly responsible for propulsion and support/control, respectively . In this scenario, the right lower extremity is possibly less tightly controlled, so that the harness effect is magnified on it. However, the size of the harness effect on knee variability (η2 = 0.06) was small, questioning the real meaning of this effect in terms of walking functionality. Third, the higher LyE and SampEn values noticed at the ankle indicated a declined (more random-like) control of this joint when walking with the harness. The increased LyE also indicated a greater local instability of this joint . Therefore, there is plausibly a negative impact of the harness on the neuromuscular control of the ankle, especially in view of the large harness effect sizes on the LyE and the SampEn (η2 = 0.31 and 0.14, respectively). Change in control was restricted to the ankle possibly because the greater inertias of the proximal joints attenuated the effect of the perturbation (here, arising from the harness) on their motions, so that their regularity and local stability remained unchanged .
In summary, it is reasonable to conclude that securing a subject in a harness during walking subtly affects lower extremity kinematics, with more important changes at the ankle. In cases where differences in gait control would be expressed through modifications in lower extremity kinematics, such differences may be masked or reinforced by having subjects walk with a harness.
Owings TM, Grabiner MD: Measuring step kinematic variability on an instrumented treadmill: how many steps are enough? J Biomech 2003, 36: 1215-1218. 10.1016/S0021-9290(03)00108-8
Visintin M, Barbeau H, Korner-Bitensky N, Mayo NE: A new approach to retrain gait in stroke patients through body weight support and treadmill stimulation. Stroke 1998, 29: 1122-1128. 10.1161/01.STR.29.6.1122
Riley PO, Paolini G, Della Croce U, Paylo K, Kerrigan DC: A kinematic and kinetic comparison of overground and treadmill walking in healthy subjects. Gait Posture 2007, 26: 17-24. 10.1016/j.gaitpost.2006.07.003
Dingwell JB, Cusumano JP, Cavanagh PR, Sternad D: Local dynamic stability versus kinematic variability of continuous overground and treadmill walking. J Biomech Eng 2001, 123: 27-32. 10.1115/1.1336798
Terrier P, Dériaz O: Kinematic variability, fractal dynamics and local dynamic stability of treadmill walking. J Neuroeng Rehabil 2011, 24: 8-12.
Threlkeld AJ, Cooper LD, Monger BP, Craven AN, Haupt HG: Temporospatial and kinematic gait alterations during treadmill walking with body weight suspension. Gait Posture 2003, 17: 235-245. 10.1016/S0966-6362(02)00105-4
Ivanenko YP, Grasso R, Macellari V, Lacquaniti F: Control of foot trajectory in human motion: role of ground contact forces in simulated reduced gravity. J Neurophysiol 2002, 87: 3070-3089.
Ivanenko YP, Grasso R, Lacquaniti F: Influence of leg muscle vibration on human walking. J Neurophysiol 2000, 84: 1737-1747.
Aaslund MK, Moe-Nilssen R: Treadmill walking with body weight support: Effect of treadmill, harness and body weight support systems. Gait Posture 2008, 28: 303-308. 10.1016/j.gaitpost.2008.01.011
Nigg BM, Cole GK, Nachbauer W: Effects of arch height of the foot on angular motion of the lower extremities in running. J Biomech 1993, 26: 909-916. 10.1016/0021-9290(93)90053-H
Vaughan CL, Davis BL, O'Connor JC: Dynamics of human gait. Cape Town, South Africa: Kiboho Publishers; 1999.
Jordan K, Challis JH, Newell KM: Walking speed influences on gait cycle variability. Gait Posture 2007, 26: 128-134. 10.1016/j.gaitpost.2006.08.010
Cappozzo A, Catani F, Leardini A, Benedetti MG, Della Croce U: Position and orientation in space of bones during movement: experimental artefacts. Clin Biomech 1996, 11: 90-100. 10.1016/0268-0033(95)00046-1
Wolf A, Swift JB, Swinney HL, Vastano JA: Determining Lyapunov exponents from a time series. Physica D 1985, 16: 285-317. 10.1016/0167-2789(85)90011-9
Cignetti F, Decker LM, Stergiou N: Sensitivity of the Wolf's and Rosenstein's algorithms to evaluate local dynamic stability from small gait data sets. Ann Biomed Eng 2011. DOI: 10.1007/s10439-011-0474-3
Ramdani S, Bouchara F, Lagarde J: Influence of noise on the sample entropy algorithm. Chaos 2009, 19: 013123. 10.1063/1.3081406
Smith B, Roan M, Lee M: The effect of evenly distributed load carrying on lower body gait dynamics for normal weight and overweight subjects. Gait Posture 2010, 32: 176-180. 10.1016/j.gaitpost.2010.04.007
LaFiandra M, Wagenaar RC, Holt KG, Obusek JP: How do load carriage and walking speed influence trunk coordination and stride parameters? J Biomech 2003, 36: 87-95. 10.1016/S0021-9290(02)00243-9
Wagenaar RC, Beek WJ: Hemiplegic gait: a kinematic analysis using walking speed as a basis. J Biomech 1992, 25: 1007-1015. 10.1016/0021-9290(92)90036-Z
Wagenaar RC, van Emmerik RE: Resonant frequencies of arms and legs identify different walking patterns. J Biomech 2000, 33: 853-861. 10.1016/S0021-9290(00)00020-8
Hausdorff JM, Rios DA, Edelberg HK: Gait variability and fall risk in community-living older adults: a 1-year prospective study. Arch Phys Med Rehabil 2001, 82: 1050-1056. 10.1053/apmr.2001.24893
Kang HG, Dingwell JB: Separating the effects of age and walking speed on gait variability. Gait Posture 2008, 27: 572-577. 10.1016/j.gaitpost.2007.07.009
Sadeghi H, Allard P, Prince F, Labelle H: Symmetry and limb dominance in able-bodied gait: a review. Gait Posture 2000, 12: 34-45. 10.1016/S0966-6362(00)00070-9
Yogev G, Plotnik M, Peretz C, Giladi N, Hausdorff JM: Gait asymmetry in patients with Parkinson's disease and elderly fallers: when does the bilateral coordination of gait require attention? Exp Brain Res 2007, 177: 336-346. 10.1007/s00221-006-0676-3
Kang HG, Dingwell JB: Dynamic stability of superior vs. inferior segments during walking in young and older adults. Gait Posture 2009, 30: 260-263. 10.1016/j.gaitpost.2009.05.003
LD is supported by the National Institutes of Health (1K99AG033684). FC and NS are supported by the National Institute on Disability and Rehabilitation Research (H133G080023). The authors thank Maris L. Hokes for her assistance in collecting data.
The authors declare that they have no competing interests.
LD designed the study and carried out the experiment. Both LD and FC analyzed the data, interpreted the results, drafted and revised the manuscript. FC conceived all figures. NS assisted in drafting and revising the manuscript. All authors approved the final version of the manuscript.
About this article
Cite this article
Decker, L.M., Cignetti, F. & Stergiou, N. Wearing a safety harness during treadmill walking influences lower extremity kinematics mainly through changes in ankle regularity and local stability. J NeuroEngineering Rehabil 9, 8 (2012). https://doi.org/10.1186/1743-0003-9-8
- Joint Angle
- Gait Cycle
- Gait Pattern
- Large Lyapunov Exponent
- Treadmill Walking