- Open Access
Prosthetic finger phalanges with lifelike skin compliance for low-force social touching interactions
© Cabibihan et al; licensee BioMed Central Ltd. 2011
- Received: 17 June 2010
- Accepted: 30 March 2011
- Published: 30 March 2011
Prosthetic arms and hands that can be controlled by the user's electromyography (EMG) signals are emerging. Eventually, these advanced prosthetic devices will be expected to touch and be touched by other people. As realistic as they may look, the currently available prosthetic hands have physical properties that are still far from the characteristics of human skins because they are much stiffer. In this paper, different configurations of synthetic finger phalanges have been investigated for their skin compliance behaviour and have been compared with the phalanges of the human fingers and a phalanx from a commercially available prosthetic hand.
Handshake tests were performed to identify which areas on the human hand experience high contact forces. After these areas were determined, experiments were done on selected areas using an indenting probe to obtain the force-displacement curves. Finite element simulations were used to compare the force-displacement results of the synthetic finger phalanx designs with that of the experimental results from the human and prosthetic finger phalanges. The simulation models were used to investigate the effects of (a) varying the internal topology of the finger phalanx and (b) varying different materials for the internal and external layers.
Results and Conclusions
During handshake, the high magnitudes of contact forces were observed at the areas where the full grasping enclosure of the other person's hand can be achieved. From these areas, the middle phalanges of the (a) little, (b) ring, and (c) middle fingers were selected. The indentation experiments on these areas showed that a 2 N force corresponds to skin tissue displacements of more than 2 mm. The results from the simulation model show that introducing an open pocket with 2 mm height on the internal structure of synthetic finger phalanges increased the skin compliance of the silicone material to 235% and the polyurethane material to 436%, as compared to a configuration with a solid internal geometry. In addition, the study shows that an indentation of 2 N force on the synthetic skin with an open pocket can also achieve a displacement of more than 2 mm, while the finger phalanx from a commercially available prosthetic hand can only achieve 0.2 mm.
- Strain Energy Function
- Indentation Experiment
- Prosthetic Hand
- Middle Phalange
- Polyurethane Material
It is not very often that our days pass by without any of the following touch-related experience: a handshake from a colleague, a caress from a spouse, a hug, a pat on the back or high fives with pals. As social beings, the need to touch and be touched is inherent among us. Indeed, it is through touch that distinct emotions such as anger, fear, disgust, love, gratitude and sympathy can be communicated . However, these typical exchanges of social touching may be limited for those who have lost their hands due to an accident, disease or war.
Prosthetic arms and hands that can be controlled by the electromyography (EMG) signals are emerging [2, 3]. Kuiken et al  demonstrated a surgical technique called targeted muscle reinnervation that transfers residual arm nerves to alternative muscle sites. They have demonstrated that this technique enables the subjects to perform real-time control of motorized shoulders, elbows, wrists and hands for grasping soft and brittle objects, like grapes or eggs, with enough force to hold them without crushing them. On the commercial front, Touch Bionics' lightweight and fully articulated prosthetic hand, the i-LIMB Hand , can now enable the amputees to perform gestures and various gripping movements (e.g. power, precision, index point, etc).
Eventually, these advanced prosthetic devices will be expected to perform affective touches on other people. This paper focuses on replicating the softness of the skin tissue on the human hand during social touching interactions. Particularly, different configurations of synthetic finger phalanges were investigated with the goal to replicate the softness of the human finger phalanges that is felt during handshakes and light touches. Here, softness is defined as the perceptual correlate of skin compliance or the amount of deformation caused by an applied force .
Aside from the functional limitations, the loss of hands implies negative psychological and emotional consequences on the person and to those around him or her. Using focus group methodology on 14 participants who have lost a limb, Gallagher and MacLachlan  found that self-image, social, physical and practical concerns are some of the key factors that are important in the adjustment process. They revealed that a common response in seeing the prosthesis for the first time was "extreme shock and disappointment". On the question of "What do you find most difficult to deal with having an artificial limb?" some participants expressed concern about the impression they made on others. One woman described that even in her own home she found it impossible for her to relax without her artificial limb on, anticipating that an unexpected visitor may arrive. Gallagher and MacLachlan noted that the common sentiment of the participants is to appear and be 'normal' again. With regard to social interactions, many of the participants encountered uncomfortable situations, wherein they stated that other people's reactions varied from patronization to complete shock.
Under similar circumstances and with limited alternatives for replacement limbs, it is understandable that depression can be one of the conditions associated with limb loss. Gallagher and MacLachlan  found that the rates of clinical depression range from 21 to 35 percent. Rybarczyk et al  highlighted five key issues that clinicians should attend to: 1) amputation is a diverse disability; 2) discrimination by others; 3) self- stigma; 4) feeling vulnerable to victimization; and 5) the role of values, meaning and perspective in positive adjustment. Using semi-structured email and face-to-face interviews on 35 prosthesis users, Murray  concluded that prosthesis use plays a social role in the lives of people with limb loss or absence. He found that the ability to conceal such use enabled participants to ward off social stigmatisation that could result into better integration into the society and the reduction of emotional problems.
Given the many issues surrounding limb loss or absence, many researchers have been addressing the technical issues in sensing, control and functionality of prosthetic arms and hands with some success (e.g. [11–16]). In terms of appearance, the current upper limb prosthetics can now be created to be indistinguishable from the natural ones with the accurate mimicry of the skin tone, hairs and pores (e.g. [17, 18]). However, these are still stiff when touched. The availability of artificial hands with lifelike appearance but not lifelike softness poses a perception problem to the person that these hands will touch. As such, finding synthetic materials that have similar properties to that of the human skin becomes important [19–21].
Among the social touches mentioned, the handshake remains one of the more acceptable gestures in many cultures. We conducted experiments to determine the areas of frequent contact during handshakes and the corresponding forces. After determining these areas, another set of experiments were done to indent a probe on the hands to obtain the amount of displacement that results from the application of forces during handshake contact. We then used Finite Element (FE) simulations to find synthetic skin designs that have similar force-displacement (i.e. skin compliance) characteristics to the human hand. The model was used to investigate the effects of (i) varying the internal topology of the finger phalanx and (ii) varying different materials for the internal and external layers. Comparisons were done on the skin compliance behaviours of the finger phalanges of the human hand, the phalanx of a commercially available prosthetic hand and the synthetic skin design with open pockets.
Handshake experiments were performed to investigate the typical range of contact forces and to find out which areas on the human hand have high contact forces during handshake. The experiments were limited to male-to-male handshake in light of the earlier findings in  that handshakes are more frequent in male-to-male pairs than in female-to-female or male-to-female pairs. Tactile force sensors were attached to the right hand of the male experimenter before he performed a series of handshakes with the test subjects. The force sensors (FingerTPS, Pressure Profile Systems, USA) are made of tactile pads that detect changes in capacitance when the upper and lower tactile pads come into contact.
Data for the Experimenter and Subjects (n = 30)
Subjects (Mean ± Std Dev)
26.35 ± 3.36
173.42 ± 4.37
65.65 ± 8.57
Hand Length (mm)
187.50 ± 11.37
Hand Width (mm)
89.53 ± 6.80
Indentation experiments on the human and prosthetic hands
Subjects' Data for the Indentation Experiment (n = 10)
Subjects (Mean ±Std Dev)
26.11 ± 3.28
173.40 ± 4.93
68.20 ± 18.23
Hand Length (mm)
188.10 ± 6.87
Hand Width (mm)
85.01 ± 8.49
Finite element modelling
This section is composed of three sub-sections that describe the synthetic skin samples, the constitutive equations and the numerical simulations.
Synthetic skin samples
The skin materials for prosthetic and robotic fingers in [11, 24] were selected for this paper. Samples of silicone (GLS 40, Prochima, s.n.c., Italy) and polyurethane (Poly 74-45, Polytek Devt Corp, USA) were previously characterized in  for their viscoelastic and hyperelastic behaviours. The silicone sample has a Shore A value of 11 while the polyurethane sample has a value of 45. A lower value indicates a low resistance to an indenter in a standard durometer test. From the durometer values above, the selected silicone material is softer as compared to the polyurethane material.
where μ i denote the shear moduli, α i are dimensionless material parameters, λ i are the principal stretch ratios, J = λ1λ2λ3 is the volume ratio and N is the number of terms used in the strain energy function. The coefficient β determines the degrees of compressibility in the energy function. The relationship of β to the Poisson's ratio, υ, is β = υ/(1-2υ).
where F is the deformation gradient and C is the right Cauchy-Green deformation tensors.
where g i is the shear relaxation modulus ratio, τ i is the relaxation time, and NG denotes the number of terms used in the relaxation function. The detailed information on how the governing equations are numerically solved have been described in the Abaqus/CAE Theory Manual .
Coefficients for the Synthetic Materials
Silicone (υ = 0.49)
τ i (sec)
μ i (MPa)
Polyurethane (υ = 0.47)
τ i (sec)
μ i (MPa)
The FE Model and Numerical Simulations
The effects of the different internal topologies are to be investigated with the use of the geometries given in Figure 3. To investigate the effects of the different material combinations, the material coefficients (i.e. data in Table 3) of the external and internal layers were set in the Abaqus™ software. For example, to have a homogeneous solid material of silicone, the inner and outer layers were given the same set of material coefficients; to have silicone as the inner layer and polyurethane as the outer layer, the material coefficients were set accordingly.
Three sets of contact interactions were specified in the model. First, a 'normal' contact behaviour was applied on the surface of the indenting plate and the external surface of the finger phalanx model. Next, a tie-connection was assumed for the 0.8 mm external layer and the rest of the finger phalanx model. Lastly, a 'normal' contact behaviour was similarly applied on the upper and lower surfaces of the 1 mm and 2 mm pocket designs as they come into contact due to the indenting plate.
The Abaqus™ 6.8-EF tetrahedral elements were used in conjunction with its automatic seed mesh feature. There were 1260 elements automatically generated for the solid internal geometry, 4838 elements for the geometry with 1 mm pocket and 3290 elements for the geometry with 2 mm pocket. The base of the finger geometry was constrained in all degrees of freedom to represent the bone structure of the human finger.
A displacement loading condition was applied on the rigid analytical surface that progressively indented each of the finger phalanx designs. The loading rate was 0.5 mm/sec. The results corresponding to the normal force (i.e. RF2) and the vertical displacement (i.e. U2) were obtained. These results will be compared to the skin compliance data of the human finger phalanges and the prosthetic hand that were obtained from the indentation experiments.
Indentation experiments on the human hand
Figure 5d shows the displacements of the finger phalanges of the 10 subjects at 2 N load under the curled and flat orientations. Paired-samples t-tests were conducted on the displacement data of each of these flat and curled pairs to evaluate the differences in their skin compliance results. There was no statistically significant difference found in the displacement data of the flat Little2 (M = 3.2631, SD = 0.8426) and curled Little2 (M = 3.7862, SD = 1.1647), t(9) = 1.6224, p = 0.1392 (two-tailed). The mean difference in the displacement data is 0.5231 with a 95% confidence interval ranging from -0.2063 to 1.2527.
No statistically significant difference was observed in the displacement data of the flat Ring2 (M = 3.7163, SD = 0.4980) and curled Ring2 (M = 3.9892, SD = 1.3457), t(9) = 0.7246, p = 0.4871 (two-tailed). The mean difference in the displacement data is 0.2729 with a 95% confidence interval ranging from -0.5789 to 1.1245.
Lastly, there was also no statistically significant difference in the displacement data in flat Middle2 (M = 3.7452, SD = 0.6623) and curled Middle2 (M = 3.5830, SD = 0.7893), t(9) = -0.6169, p = 0.5526 (two-tailed). The mean difference in the displacement data is 0.1622 with a 95% confidence interval ranging from -0.7573 to 0.4328. In summary, these data show that there are no significant differences in the skin compliance of the Little2, Ring2 and Middle2 in flat and curled orientations.
Effect of the open pockets
Effect of varying the layers
The human skin is tough, compliant and has self-healing properties. Technologies that can replicate all these properties are not yet available for prosthetic skins. Therefore, it is important to investigate the effects of a two-layered synthetic skin, which can give insights on how to satisfy the requirements for softness for social touching interactions and other requirements for wear, puncture and tear.
This section describes the effects of having a 0.8 mm outer layer of one type of material and an internal layer of another material. The results are shown as the thinner line types in Figure 7 (i.e. SIL Inner PU Outer and PU Inner SIL Outer). The first four curves clustered at the bottom part of the figure are the simulation results from the solid internal geometry configuration. A 2 N compressive load for a combination of PU inner layer and an outer layer of SIL resulted into a displacement of 0.44 mm, or a 4.8% increase from a homogeneous PU material condition. With a combined inner layer of SIL and outer layer of PU, the displacement was 0.72 mm or a 6.5% decrease from a 'SIL-only' material condition. These results were expected because the SIL material has a lower durometer value (i.e. softer) as compared to the PU material. For the remaining combinations, the changes in the displacements at the 2 N compressive load correspond to an increase or decrease of displacement values to within 7% from the homogeneous material condition.
The significant effect of having the two layers of materials can be observed during the unsupported deflection of the upper part of the pocket, where the effects of the material softness come in. Looking at the results of the internal pockets with 2 mm heights, we can observe that at the 2 mm displacement the magnitude of the force for the 'SIL-only' condition is 0.39 N and it is 0.5 N for the 'SIL inner and PU outer' configuration; or a 28% increase in force value. For the 'PU-only' configuration, the force is reduced from 0.84 N to 0.72 N for the 'PU inner and SIL outer' configuration, or about 14% decrease.
Alternatively, we can analyze the effects of having the two material layers by comparing the slopes (i.e. Δdisplacement/Δforce) of the rising part of the curves for the 1 mm and 2 mm internal pockets in Figure 7. Again, for the geometries with the 2 mm height pockets as an example, the rising slope is 6.64 for the 'SIL-only' and 5.03 for 'SIL inner and PU outer' conditions. The slope is 3.25 for 'PU-only' and 4.29 for 'PU inner and SIL outer' conditions. Taken together, the 0.8 mm outer layer significantly affects the slope of the rising curve, which results into an increase or decrease of the force magnitude. These occur before the top layer of the pocket comes into contact with the bottom layer of the pocket and eventually stiffens.
Comparisons of simulation results against the prosthetic and human finger phalanges
The figures in the right column in Figure 8 show the magnitude of the displacements corresponding to the 1 N force indentation. The results labelled with 'curled' and 'flat' in Figure 8 are from the experimental data taken from the finger phalanges indentation of the 10 test subjects. From the simulation results of the synthetic finger phalanges, it can be observed from the bars that having a 2 mm height internal pocket can introduce significant improvements in the skin compliance. Such results are important particularly when they are compared against the skin compliance of the finger phalanges of a commercially available prosthetic hand. The silicone material used by the manufacturer of the prosthetic hand was stiff as shown by the 0.2 mm deformation. Many prosthetic hands are being sold for their cosmetic appearance and durability. The development of prosthetic hands for natural social touching interactions has not been the norm.
This paper addressed the continuing need for improved methods and designs that can make prosthetic hands and arms unnoticed during social touching situations. In addition to lifelike appearance, warmth and motion, prosthetic skins that can replicate the natural softness of the human hand may be able to shield the user from social stigma. This could lead to the faster improvement of his or her emotional well-being and permit the resumption of a normal life (cf. [10, 17, 29]).
In this paper, the areas of the hand where typical contact occurs during male-to-male handshakes were determined. The results show that the following areas of the hand have contact forces greater than 2 N when grasped during handshake: (a) the palm and the back of the palm, (b) the thumb, (c) the proximal phalanx of the little finger and the (e) middle phalanges of the little, (f) ring, and (g) middle fingers. These are the areas that envelope the handshaking partner's hand for a full grasp.
The middle phalanges of the little, ring and middle fingers were selected for indentations with a testing machine. The force-displacement curves were obtained on both the flat hand position, which represents tapping or caressing postures, and the curled-fingers position, which represents handshake postures. The indentation results show that the skin tissues at the finger phalanges are compliant and are exhibiting large displacements with minimal forces applied, i.e., a 1 N force corresponds to skin tissue displacements of more than 2 mm. The results also show that there are no significant differences in the force-displacement data on flat-hand position and curled-fingers position on the middle phalanges of the little, ring and middle fingers.
Three-dimensional finite element models were presented for investigating the effects of varying the internal topology and varying the material layers in an attempt to duplicate the skin compliance of human finger phalanges. The following conclusions can be made from the simulation results. First, the skin compliance can be increased by introducing open pockets on the internal structure of a synthetic finger phalanx. An arc-shaped pocket with a 2 mm height on the internal structure increased the skin compliance of the silicone material to as high as 235% and the polyurethane material to 436%, as compared to a configuration with a solid internal geometry.
Second, having one type of material for the 0.8 mm external layer and another type for the internal layer can affect the deflection of the finger phalanges' surface, but this combination has minimal effect when the top layer of the pocket comes into contact with the base of the finger phalanx. By knowing the effects of having multi-material layers, we can take advantage of a synthetic skin design with a stiff external layer and a soft internal layer. A stiff external layer can better protect the tactile sensors and electronics that may be embedded on the internal structure, while the soft internal layer can satisfy the requirements for more natural social touching.
Lastly, the simulation results show that the synthetic skins with the configurations described herein could achieve lifelike skin compliance for light social touches, especially under applied forces of about 1 N. The internal pockets can significantly improve the compliance of the synthetic skins that will be used for prosthetics. Future studies can investigate other softer materials (i.e. materials with lower Shore durometer values), find the optimal thickness for the internal and external layers and the internal pockets, and optimize the right combinations of materials to be used as the internal and the external layer.
This work was supported by the project 'Design of Prosthetic Skins with Humanlike Softness' (R-263-000-506-133) funded by the Academic Research Fund, Ministry of Education, Singapore. We thank Lifeforce Limbs and Rehab Pte Ltd for the prosthetic hand sample.
- Hertenstein MJ, Keltner D, App B, Bulleit BA, Jaskolka AR: Touch communicates distinct emotions. Emotion 2006, 6: 528-533. 10.1037/1528-35126.96.36.1998View ArticlePubMedGoogle Scholar
- Kuiken TA, Dumanian GA, Lipschutz RD, Miller LA, Stubblefield KA: The use of targeted muscle reinnervation for improved myoelectric prosthesis control in a bilateral shoulder disarticulation amputee. Prosthetics and Orthotics International 2004, 28: 245-253.PubMedGoogle Scholar
- Kuiken TA, Miller LA, Lipschutz RD, Lock BA, Stubblefield K, Marasco PD, Zhou P, Dumanian GA: Targeted reinnervation for enhanced prosthetic arm function in a woman with a proximal amputation: a case study. Lancet 2007, 369: 371-380. 10.1016/S0140-6736(07)60193-7View ArticlePubMedGoogle Scholar
- Kuiken TA, Li G, Lock BA, RD L, Miller LA, Stubblefield KA, Englehart KB: Targeted muscle reinnervation for real-time myoelectric control of multifunction artificial arms. Journal of the American Medical Association 2009, 301: 619-628. 10.1001/jama.2009.116PubMed CentralView ArticlePubMedGoogle Scholar
- The i-Limb Hand[http://www.touchbionics.com/i-LIMB]
- Friedman RM, Hester KD, Green BG, LaMotte RH: Magnitude estimation of softness. Exp Brain Res 2008,191(2):133-42. 10.1007/s00221-008-1507-5PubMed CentralView ArticlePubMedGoogle Scholar
- Gallagher P, MacLachlan M: Adjustment to an artificial limb: A qualitative perspective. Journal of Health Psychology 2001, 6: 85-100. 10.1177/135910530100600107View ArticlePubMedGoogle Scholar
- Gallagher P, MacLachlan M: Development and psychometric evaluation of the trinity amputation and prosthesis experience scales (TAPES). Rehabilitation Psychology 2000, 45: 130-154. 10.1037/0090-55188.8.131.52View ArticleGoogle Scholar
- Rybarczyk B, Nicholas JJ, Nyenhuis DL: Coping with a leg amputation: Integrating research and clinical practice. Rehabilitation Psychology 1997, 42: 241-255. 10.1037/0090-55184.108.40.206View ArticleGoogle Scholar
- Murray CD: The social meanings of prosthesis use. Journal of Health Psychology 2005, 10: 425-441. 10.1177/1359105305051431View ArticlePubMedGoogle Scholar
- Edin BB, Ascari L, Beccai L, Roccella S, Cabibihan JJ, Carrozza MC: Bio-inspired sensorization of a biomechatronic robot hand for the grasp-and-lift task. Brain Research Bulletin 2008, 75: 785-795. 10.1016/j.brainresbull.2008.01.017View ArticlePubMedGoogle Scholar
- Matrone G, Cipriani C, Secco E, Magenes G, Carrozza M: Principal components analysis based control of a multi-dof underactuated prosthetic hand. Journal of NeuroEngineering and Rehabilitation 2010, 7: 16. 10.1186/1743-0003-7-16PubMed CentralView ArticlePubMedGoogle Scholar
- Abboudi RL, Glass CA, Newby NA, Flint JA, Craelius W: A biomimetic controller for a multifinger prosthesis. IEEE Transactions on Rehabilitation Engineering 1999, 7: 121-129. 10.1109/86.769401View ArticlePubMedGoogle Scholar
- Carrozza MC, Suppo C, Sebastiani F, Massa B, Vecchi F, Lazzarini R, Cutkosky MR, Dario P: The SPRING hand: Development of a self-adaptive prosthesis for restoring natural grasping. Autonomous Robots 2004, 16: 125-141. 10.1023/B:AURO.0000016863.48502.98View ArticleGoogle Scholar
- Castellini C, Fiorilla AE, Sandini G: Multi-subject/daily-life activity EMG-based control of mechanical hands. Journal of NeuroEngineering and Rehabilitation 2009., 6: 10.1186/1743-0003-6-41Google Scholar
- Carrozza MC, Cappiello G, Micera S, Edin BB, Beccai L, Cipriani C: Design of a cybernetic hand for perception and action. Biological Cybernetics 2006, 95: 629-644. 10.1007/s00422-006-0124-2PubMed CentralView ArticlePubMedGoogle Scholar
- Leow MEL, Pho RWH, Pereira BP: Esthetic prostheses in minor and major upper limb amputations. Hand Clinics 2001, 17: 489-497.PubMedGoogle Scholar
- Cabibihan JJ, Pattofatto S, Jomaa M, Benallal A, Carrozza MC: Towards humanlike social touch for sociable robotics and prosthetics: Comparisons on the compliance, conformance and hysteresis of synthetic and human fingertip skins. International Journal of Social Robotics 2009, 1: 29-40. 10.1007/s12369-008-0008-9View ArticleGoogle Scholar
- Cabibihan JJ, Pradipta R, Chew YZ, Ge SS: Towards humanlike social touch for prosthetics and sociable robotics: Handshake experiments and finger phalange indentations. In Advances in Robotics. Volume 5744. Edited by: Kim JH, Ge SS, Vadakkepat P, Jesse N. Springer; 2009:73-79. LNCS full_textView ArticleGoogle Scholar
- Cabibihan JJ, Ge SS: Towards humanlike social touch for prosthetics and sociable robotics: Three-dimensional finite element simulations of synthetic finger phalanges. In Advances in Robotics. Volume 5744. Edited by: Kim JH, Ge SS, Vadakkepat P, Jesse N. Springer; 2009:80-86. LNCS full_textView ArticleGoogle Scholar
- Greenbaum PE, Rosenfeld HM: Varieties of touching in greetings: Sequential structure and sex-related differences. Journal of Nonverbal Behavior 1980, 5: 13-25. 10.1007/BF00987051View ArticleGoogle Scholar
- Chaplin WF, Phillips JB, Brown JD, Clanton NR, Stein JL: Handshaking, gender, personality, and first impressions. Journal of Personality and Social Psychology 2000, 79: 110-117. 10.1037/0022-35220.127.116.11View ArticlePubMedGoogle Scholar
- Beccai L, Roccella S, Ascari L, Valdastri P, Sieber A, Carrozza MC, Dario P: Development and experimental analysis of a soft compliant tactile microsensor for anthropomorphic artificial hand. Mechatronics, IEEE/ASME Transactions on 2008, 13: 158-168. 10.1109/TMECH.2008.918483View ArticleGoogle Scholar
- Storakers B: On material representation and constitutive branching in finite compressible elasticity. Journal of the Mechanics and Physics of Solids 1986, 34: 125-145. 10.1016/0022-5096(86)90033-5View ArticleGoogle Scholar
- Wu JZ, Dong RG, Rakheja S, Schopper AW, Smutz WP: A structural fingertip model for simulating of the biomechanics of tactile sensation. Medical Engineering and Physics 2004, 26: 165-175. 10.1016/j.medengphy.2003.09.004View ArticlePubMedGoogle Scholar
- Tschoegl NW: The phenomenological theory of linear viscoelastic behavior: An introduction. New York: Springer-Verlag; 1989.View ArticleGoogle Scholar
- Dassault: Abaqus Theory Manual v.6.8-EF. Providence, RI, USA: Dassault Systemes Simulia Corp; 2008.Google Scholar
- Pillet J, Didierjean-Pillet A: Aesthetic hand prosthesis: gadget or therapy? Presentation of a new classification. The Journal of Hand Surgery 2001, 28: 523-528.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.